Main Error Enumeration
ID
 Error
 Description
 
 0
 No error
 -
 
 1
 Syntax Error
 There is a wrong set of parameters or parameter ranges.
 
 2
 Hardware Error
 Steps lost on one or more hardware components, or component not initialized or not functioning.
 
 3
 Not Executed Error
 There was an error in previous part command.
 
 4
 Clot Error
 Blood clot detected.
 
 5
 Barcode Error
 Barcode could not be read or is missing.
 
 6
 Insufficient Liquid Error
 Not enough liquid available.
 
 7
 Tip Present Error
 A tip has already been picked up.
 
 8
 No Tip Error
 Tip is missing or not picked up.
 
 9
 No Carrier Error
 No carrier present for loading.
 
 10
 Execution Error
 A step or a part of a step could not be processed.
 
 11
 Pressure LLD Error
 A dispense with pressure liquid level detection is not allowed.
 
 12
 Calibrate Error
 No capacitive signal detected during carrier calibration procedure.
 
 13
 Unload Error
 Not possible to unload the carrier due to occupied loading tray position.
 
 14
 Pressure LLD Error
 Pressure liquid level detection in a consecutive aspiration is not allowed.
 
 15
 Parameter Error
 Dispense in jet mode with pressure liquid level detection is not allowed.
 
 16
 Cover Open Error
 Cover not closed or can not be locked.
 
 17
 Improper Aspiration Error

Improper Dispense Error
 The pressure-based aspiration / dispensation control reported an error ( not enough liquid ).
 
 18
 Wash Liquid Error
 Waste full or no more wash liquid available.
 
 19
 Temperature Error
 Incubator temperature out of range.
 
 20
 TADM overshot
 Overshot of limits during aspirate or dispense.

Note:

On aspirate this error is returned as main error 17.

On dispense this error is returned as main error 4.
 
 21
 Labware Error
 Labware not available.
 
 22
 Labware Gripped Error
 Labware already gripped.
 
 23
 Labware Lost Error
 Labware lost during transport.
 
 24
 Illegal target plate position
 Cannot place plate, plate was gripped in a wrong direction.
 
 25
 Illegal Intervention Error
 Cover was opened or a carrier was removed manually.
 
 26
 TADM undershot
 Undershot of limits during aspirate or dispense.

Note:

On aspirate this error is returned as main error 4.

On dispense this error is returned as main error 17.
 
 27
 Position Error
 The position is out of range.
 
28
 Unexpected cLLD Error
 The cLLD detected a liquid level above start height of liquid level search.
 
29
 Area already occupied
 Instrument region already reserved.
 
30
 Impossible to occupy area
 A region on the instrument cannot be reserved.
 
31
 Anti drop control error
 Anti drop controlling out of tolerance.
 
32
 Decapper error
 Decapper lock error while screw / unscrew a cap by twister channels.
 
33
 Decapper handling error
 Decapper station error while lock / unlock a cap.
 
99
 Slave Error
 Slave error.
 
 100
 Wrong Carrier Error
 Wrong carrier barcode detected.
 
 101
 No Carrier Barcode Error
 Carrier barcode could not be read or is missing.
 
 102
 Liquid Level Error
 Liquid surface not detected.

This error is created from main / slave error 06/70, 06/73 and 06/87.
 
 103
 Not Detected Error
 Carrier not detected at deck end position.
 
 104
 Not Aspirated Error
 Dispense volume exceeds the aspirated volume.

This error is created from main / slave error 02/54.
 
105
 Improper Dispensation Error
 The dispensed volume is out of tolerance (may only occur for Nano Pipettor Dispense steps). 

This error is created from main / slave error 02/52 and 02/54.
 
106
 No Labware Error
 The labware to be loaded was not detected by autoload module.

Note:

May only occur on a Reload Carrier step if the labware property 'MlStarCarPosAreRecognizable' is set to 1.
 
107
 Unexpected Labware Error
 The labware contains unexpected barcode ( may only occur on a Reload Carrier step ).
 
108
 Wrong Labware Error
 The labware to be reloaded contains wrong barcode ( may only occur on a Reload Carrier step ).
 
109
 Barcode Mask Error
 The barcode read doesn't match the barcode mask defined.
 
110
 Barcode Not Unique Error
 The barcode read is not unique. Previously loaded labware with same barcode was loaded without unique barcode check.
 
111
 Barcode Already Used Error
 The barcode read is already loaded as unique barcode ( it's not possible to load the same barcode twice ).
 
112
 Kit Lot Expired Error
 Kit Lot expired.
 
113
 Delimiter Error
 Barcode contains character which is used as delimiter in result string.
 
