<?xml version="1.0" encoding="UTF-8" standalone="yes"?>
<!-- edited with XMLSpy v2009 sp1 (http://www.altova.com) by USER USER (CEA - CENTRE DE CADARACHE) -->
<?modxslt-stylesheet type="text/xsl" media="fuffa, screen and $GET[stylesheet]" href="./%24GET%5Bstylesheet%5D" alternate="no" title="Translation using provided stylesheet" charset="UTF-8" ?>
<?modxslt-stylesheet type="text/xsl" media="screen" alternate="no" title="Show raw source of the XML file" charset="UTF-8" ?>
<!--
  -* Generated by mod-xslt 1.3.8; http://www.mod-xslt2.com/
  -* Copyright (C) 2002,2003 Carlo Contavalli - <ccontavalli at masobit.net>
  -* derived from work by Philipp Dunkel and others (http://www.mod-xslt2.com/main/credits.xml)
  -* Thanks to http://www.masobit.net/ for paying me while working on mod-xslt
  -* and for providing resources to the project. -->
<xs:schema xmlns:xs="http://www.w3.org/2001/XMLSchema" elementFormDefault="qualified" attributeFormDefault="unqualified">
	<xs:include schemaLocation="../utilities/dd_support.xsd"/>
	<xs:complexType name="thomson_scattering_channel">
		<xs:annotation>
			<xs:documentation>Thomson scattering channel</xs:documentation>
		</xs:annotation>
		<xs:sequence>
			<xs:element name="name">
				<xs:annotation>
					<xs:documentation>Name of the channel</xs:documentation>
					<xs:appinfo>
						<type>static</type>
					</xs:appinfo>
				</xs:annotation>
				<xs:complexType>
					<xs:group ref="STR_0D"/>
				</xs:complexType>
			</xs:element>
			<xs:element name="identifier">
				<xs:annotation>
					<xs:documentation>ID of the channel</xs:documentation>
					<xs:appinfo>
						<type>static</type>
					</xs:appinfo>
				</xs:annotation>
				<xs:complexType>
					<xs:group ref="STR_0D"/>
				</xs:complexType>
			</xs:element>
			<xs:element name="position" type="rzphi0d_static">
				<xs:annotation>
					<xs:documentation>Average position of the measurements (intersection between laser beam and line of sight)</xs:documentation>
				</xs:annotation>
			</xs:element>
			<xs:element name="delta_position" type="rzphi1d_dynamic_aos1_common_time">
				<xs:annotation>
					<xs:documentation>Incremental variation of the position of the measurements, due to e.g. different lasers not intersecting the line of sight at the same position. The actual position is then the static position + delta_position</xs:documentation>
				</xs:annotation>
			</xs:element>
			<xs:element name="distance_separatrix_midplane" type="signal_flt_1d">
				<xs:annotation>
					<xs:documentation>Distance between the measurement position and the separatrix, mapped along flux surfaces to the outboard midplane, in the major radius direction. Positive value means the measurement is outside of the separatrix.</xs:documentation>
					<xs:appinfo>
						<units>m</units>
						<introduced_after_version>3.32.1</introduced_after_version>
					</xs:appinfo>
				</xs:annotation>
			</xs:element>
			<xs:element name="t_e" type="signal_flt_1d">
				<xs:annotation>
					<xs:documentation>Electron temperature</xs:documentation>
					<xs:appinfo>
						<units>eV</units>
					</xs:appinfo>
				</xs:annotation>
			</xs:element>
			<xs:element name="n_e" type="signal_flt_1d">
				<xs:annotation>
					<xs:documentation>Electron density</xs:documentation>
					<xs:appinfo>
						<units>m^-3</units>
					</xs:appinfo>
				</xs:annotation>
			</xs:element>
		</xs:sequence>
	</xs:complexType>
	<xs:element name="thomson_scattering">
		<xs:annotation>
			<xs:documentation>Thomson scattering diagnostic</xs:documentation>
			<xs:appinfo>
				<lifecycle_status>alpha</lifecycle_status>
				<lifecycle_version>3.0.4</lifecycle_version>
				<lifecycle_last_change>3.33.0</lifecycle_last_change>
			</xs:appinfo>
		</xs:annotation>
		<xs:complexType>
			<xs:sequence>
				<xs:element ref="ids_properties"/>
				<xs:element name="equilibrium_id" type="ids_identification">
					<xs:annotation>
						<xs:documentation>ID of the IDS equilibrium used to map measurements - we may decide that this is superseeded when the systematic documentation of input provenance is adopted</xs:documentation>
						<xs:appinfo>
							<introduced_after_version>3.32.1</introduced_after_version>
						</xs:appinfo>
					</xs:annotation>
				</xs:element>
				<xs:element name="midplane" type="identifier_static">
					<xs:annotation>
						<xs:documentation>Choice of midplane definition for the mapping of measurements on an equilibrium</xs:documentation>
						<xs:appinfo>
							<doc_identifier>utilities/midplane_identifier.xml</doc_identifier>
							<introduced_after_version>3.32.1</introduced_after_version>
						</xs:appinfo>
					</xs:annotation>
				</xs:element>
				<xs:element name="channel" type="thomson_scattering_channel" maxOccurs="150">
					<xs:annotation>
						<xs:documentation>Set of channels (lines-of-sight)</xs:documentation>
						<xs:appinfo>
							<coordinate1>1...N</coordinate1>
						</xs:appinfo>
					</xs:annotation>
				</xs:element>
				<xs:element name="latency">
					<xs:annotation>
						<xs:documentation>Upper bound of the delay between physical information received by the detector and data available on the real-time (RT) network.</xs:documentation>
						<xs:appinfo>
							<type>static</type>
							<units>s</units>
							<introduced_after_version>3.32.1</introduced_after_version>
						</xs:appinfo>
					</xs:annotation>
					<xs:complexType>
						<xs:group ref="FLT_0D"/>
					</xs:complexType>
				</xs:element>
				<xs:element ref="code"/>
				<xs:element ref="time"/>
			</xs:sequence>
		</xs:complexType>
	</xs:element>
</xs:schema>
