cwEPR Info file - v. 0.1.4 (2020-01-21)

GENERAL
Filename:               glas001
Date start:             2013-04-04
Date end:               2013-04-04
Time start:             08:00:00
Time end:               08:40:00
Operator:               Max Mustermann
Label:                  leeres Roehrchen
Purpose:                Test auf Glassignal

SAMPLE
Name:                   leeres Quarzglas-Roehrchen
ID:                     42
Description:            2 mm od, 1 mm id
Solvent:                N/A
Preparation:            N/A
Tube:                   2 x 1 x 120 mm

EXPERIMENT
Type:                   field-sweep
Runs:                   1
Variable parameter:     field
Increment:              0.1 G

SPECTROMETER
Model:                  ESP380E
Software:               esp380e

MAGNETIC FIELD
Field probe type:       Hall
Field probe model:      xxx
Start:                  339.50 mT
Stop:                   349.50 mT
Step:                   100/1024
Sequence:               up
Controller:             Bruker ER 032M
Power supply:           Bruker ER 083 CS

BRIDGE
Model:                  Bruker ER 046 MRT
Controller:             Bruker EMX 048 T
Attenuation:            25 dB
Power:                  0.63 mW
Detection:              diode
Frequency counter:      HP 5352B
MW frequency:           9.66567 GHz

SIGNAL CHANNEL
Model:                  Bruker ER 023M
Accumulations:          25
Modulation frequency:   100 kHz
Modulation amplitude:   0.2 mT
Receiver gain:          5.1e5
Conversion time:        81.92 ms
Time constant:          20.48 ms
Phase:                  0 deg

DIGITAL FILTER
Mode:                   Manual
Number of Points:       0

PROBEHEAD
Type:                   dielectric
Model:                  Bruker ER 4118X-MD5
Coupling:               critical

TEMPERATURE
Temperature:            270 K
Controller:             Oxford ITC 503
Cryostat:               Oxford 935
Cryogen:                N2

FIELD CALIBRATION
Gaussmeter:
Method:                 Standard | StartEnd  % Wichtig als "Schalter" beim Parsen
Standard:               LiLiF
Signal field:           3448.99 G       % Felder fuer Freiburg
MW Frequency:           9.666147 GHz
Deviation:              0.18 G

COMMENT
Infofile changed from version 0.1, therefore some fields are missing