parameters = {"allparameters":[{"advanced":False,"available":True,"default":"7SN999999","description":"identnr.+softwareversion[+serialnr.]","fbnr":0,"group0":13,"group1":"","group2":"","highly secured":False,"longname":"Identification string","max":"","min":"","name":"identstrng","parameter":1,"poll":False,"process":0,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"primary node address: network parameter FLOW-BUS","fbnr":1,"group0":1,"group1":"","group2":"","highly secured":True,"longname":"Primary node address","max":128,"min":0,"name":"pna","parameter":2,"poll":False,"process":0,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"secundary node address: network parameter FLOW-BUS","fbnr":2,"group0":1,"group1":"","group2":"","highly secured":True,"longname":"Secondary node address","max":128,"min":0,"name":"sna","parameter":3,"poll":False,"process":0,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1","description":"next node address: network parameter FLOW-BUS","fbnr":3,"group0":1,"group1":"","group2":"","highly secured":False,"longname":"Next node address","max":128,"min":0,"name":"nna","parameter":4,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"32","description":"last node address: network parameter FLOW-BUS","fbnr":4,"group0":1,"group1":"","group2":"","highly secured":False,"longname":"Last node address","max":128,"min":0,"name":"lna","parameter":5,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"67","description":"FLOW-BUS arbitrage setting and/or automatic optimization","fbnr":5,"group0":1,"group1":"","group2":"","highly secured":True,"longname":"Arbitrage","max":255,"min":0,"name":"arbitrage","parameter":6,"poll":False,"process":0,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"","description":"init and reset security key commands for network/parameter settings","fbnr":10,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Initreset","max":255,"min":0,"name":"initreset","parameter":7,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"measured value (100% = 32000)","fbnr":0,"group0":2,"group1":"","group2":"","highly secured":False,"longname":"Measure","max":41942,"min":-23593,"name":"measure","parameter":8,"poll":True,"process":"","read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"setpoint: wanted value (100% = 32000)","fbnr":1,"group0":2,"group1":18,"group2":"","highly secured":False,"longname":"Setpoint","max":32767,"min":0,"name":"setpoint","parameter":9,"poll":True,"process":"","read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"setpoint ramp signal 0..100 % in up to slope x 0.1 sec.","fbnr":2,"group0":18,"group1":"","group2":"","highly secured":False,"longname":"Setpoint slope","max":30000,"min":0,"name":"setpslope","parameter":10,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"analog input signal, normally used for ext. setp. (100% = 32000)","fbnr":3,"group0":2,"group1":18,"group2":"","highly secured":False,"longname":"Analog input","max":41942,"min":-23593,"name":"analoginp","parameter":11,"poll":True,"process":"","read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"control mode selection for instrument or module","fbnr":4,"group0":18,"group1":"","group2":"","highly secured":False,"longname":"Control mode","max":255,"min":0,"name":"cntrlmode","parameter":12,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0.0","description":"polynomial constant A (offset)","fbnr":5,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Polynomial constant A","max":3.40282E38,"min":-3.40282E38,"name":"polycnst A","parameter":13,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1.0","description":"polynomial constant B (span]","fbnr":6,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Polynomial constant B","max":3.40282E38,"min":-3.40282E38,"name":"polycnst B","parameter":14,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0.0","description":"polynomial constant C","fbnr":7,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Polynomial constant C","max":3.40282E38,"min":-3.40282E38,"name":"polycnst C","parameter":15,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0.0","description":"polynomial constant D","fbnr":8,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Polynomial constant D","max":3.40282E38,"min":-3.40282E38,"name":"polycnst D","parameter":16,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0.0","description":"polynomial constant E (offset) for setpoint or power value","fbnr":9,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Polynomial constant E","max":3.40282E38,"min":-3.40282E38,"name":"polycnst E","parameter":17,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1.0","description":"polynomial constant F (span) for setpoint or power value","fbnr":10,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Polynomial constant F","max":3.40282E38,"min":-3.40282E38,"name":"polycnst F","parameter":18,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0.0","description":"polynomial constant G for setpoint or power value","fbnr":11,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Polynomial constant G","max":3.40282E38,"min":-3.40282E38,"name":"polycnst G","parameter":19,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0.0","description":"polynomial constant H for setpoint or power value","fbnr":12,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Polynomial constant H","max":3.40282E38,"min":-3.40282E38,"name":"polycnst H","parameter":20,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1.0","description":"readout value at 100% in capacity (readout) unit","fbnr":13,"group0":3,"group1":19,"group2":"","highly secured":False,"longname":"Capacity 100%","max":3.40282E38,"min":-3.40282E38,"name":"capacity","parameter":21,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"3","description":"sensor type information for actual reading and sensor/controller indication","fbnr":14,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Sensor type","max":255,"min":0,"name":"sensortype","parameter":22,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"pointer to capacity (readout) unit (classic unit table]","fbnr":15,"group0":3,"group1":19,"group2":"","highly secured":False,"longname":"Capacity unit index","max":255,"min":0,"name":"capunit","parameter":23,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"fluid number: pointer to fluidset (for e.g. polynome, name and cap.)","fbnr":16,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Fluidset index","max":8,"min":0,"name":"fluidnr","parameter":24,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"AIR","description":"name of fluid","fbnr":17,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Fluid name","max":"","min":"","name":"fluidname","parameter":25,"poll":False,"process":"","read":True,"secured":True,"varlength":10,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"node address of module with operation rights","fbnr":18,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Claim node","max":128,"min":0,"name":"claimnode","parameter":26,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"contains number(s) of changed par (0xXX par nr, 0xFF more than one par changed)","fbnr":19,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Modify","max":255,"min":0,"name":"modify","parameter":27,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"status information of several alarms/errors in the instrument","fbnr":20,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Alarm info","max":255,"min":0,"name":"alarminfo","parameter":28,"poll":True,"process":"","read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"32","description":"amount of channels which can be operated","fbnr":12,"group0":17,"group1":"","group2":"","highly secured":False,"longname":"Channel amount","max":120,"min":1,"name":"chanamount","parameter":29,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"first channel that can be operated","fbnr":13,"group0":17,"group1":"","group2":"","highly secured":False,"longname":"First channel","max":120,"min":1,"name":"firstchan","parameter":30,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"32","description":"last channel that can be operated","fbnr":14,"group0":17,"group1":"","group2":"","highly secured":False,"longname":"Last channel","max":120,"min":1,"name":"lastchan","parameter":31,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"operation by HOST computer enable flag","fbnr":1,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"<hostcontrl>","max":1,"min":0,"name":"hostcontrl","parameter":32,"poll":False,"process":9,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"alarm message string with unit type information","fbnr":0,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Alarm message unit type","max":"","min":"","name":"alrmmsgTA","parameter":33,"poll":True,"process":10,"read":True,"secured":False,"varlength":16,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"","description":"alarm message string with unit number information","fbnr":1,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Alarm message number","max":"","min":"","name":"alrmmsgnr","parameter":34,"poll":True,"process":10,"read":True,"secured":False,"varlength":16,"vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"","description":"status of relays/potential free contacts ('0' = not activated, '1' = activated)","fbnr":2,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Relay status","max":"","min":"","name":"relstatus","parameter":35,"poll":False,"process":10,"read":True,"secured":False,"varlength":8,"vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"actual value of counter","fbnr":0,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Actual counter value","max":3.40282E38,"min":0,"name":"actualval","parameter":36,"poll":True,"process":"","read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"","description":"signal input selection (' '=no value,'+'=pos value,'-'=neg value input)","fbnr":1,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Signal input selection","max":"","min":"","name":"signinpsel","parameter":37,"poll":False,"process":"","read":True,"secured":False,"varlength":8,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"","description":"external reset input enable/disable ('E'=enable,' '=disable)","fbnr":2,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Reset input selection","max":"","min":"","name":"resinpsel","parameter":38,"poll":False,"process":"","read":True,"secured":False,"varlength":8,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"limit/batch for counter in sensor standard units","fbnr":3,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"<limit>","max":3.40282E38,"min":0,"name":"limit","parameter":39,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"00000000","description":"delay time string in days,hours,minutes,seconds","fbnr":4,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Delay time","max":99235959,"min":0,"name":"delaytime","parameter":40,"poll":False,"process":"","read":True,"secured":False,"varlength":8,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"00000000","description":"duration time string in days,hours,minutes,seconds","fbnr":5,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Duration time","max":99235959,"min":0,"name":"duratntime","parameter":41,"poll":False,"process":"","read":True,"secured":False,"varlength":8,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"00000000","description":"valve output setting ('0'=do nothing, '1'=close valve)","fbnr":6,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Valve output setting","max":"","min":"","name":"vlvoutset","parameter":42,"poll":False,"process":"","read":True,"secured":False,"varlength":8,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"LLLLLLLL","description":"relay output setting ('L'=low, 'H'=high, 'P'=pulse (1 sec.))","fbnr":7,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Relay output setting","max":"","min":"","name":"reloutset","parameter":43,"poll":False,"process":"","read":True,"secured":False,"varlength":8,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"operation mode of T/A module","fbnr":8,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Operation mode T/A","max":9,"min":0,"name":"opermodeTA","parameter":44,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"ln/min","description":"readout unit string","fbnr":9,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Readout unit","max":"","min":"","name":"readunit","parameter":45,"poll":False,"process":"","read":True,"secured":False,"varlength":7,"vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"1","description":"readout factor matching readout unit string","fbnr":10,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Readout factor","max":10000000000,"min":1E-10,"name":"readfact","parameter":46,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"","description":"reset unit command (1=reset T/A unit)","fbnr":12,"group0":5,"group1":"","group2":"","highly secured":False,"longname":"Reset unit","max":1,"min":0,"name":"resetunit","parameter":47,"poll":False,"process":"","read":False,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0.1","description":"valve output differentiation time constant downwards","fbnr":9,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Valve differentiator down","max":3.40282E38,"min":0,"name":"TdValveDn","parameter":48,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0.1","description":"valve output differentiation time constant upwards","fbnr":10,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Valve differentiator up","max":3.40282E38,"min":0,"name":"TdValveUp","parameter":49,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"5.8","description":"sensor signal differentiation time constant downwards","fbnr":11,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Sensor differentiator down","max":3.40282E38,"min":0,"name":"TdSensorDn","parameter":50,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"5.6","description":"sensor signal differentiation time constant upwards","fbnr":12,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Sensor differentiator up","max":3.40282E38,"min":0,"name":"TdSensorUp","parameter":51,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"6","description":"cycle time * 10 msec. main loop signal processing","fbnr":12,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Cycle time","max":255,"min":0,"name":"CycleTime","parameter":52,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"f","write":True},
{"advanced":False,"available":True,"default":"0","description":"analog mode selection for analog operation","fbnr":3,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog mode","max":255,"min":0,"name":"AnalogMode","parameter":53,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"reference voltage output signal for analog operation","fbnr":6,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Reference voltage","max":65535,"min":0,"name":"VrefOutput","parameter":54,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"valve output signal (24-bit number in range 0..14.3Vdc/0..23.3Vdc)","fbnr":1,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve output","max":16777215,"min":0,"name":"ValveOut","parameter":55,"poll":True,"process":114,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0,001","description":"dynamic display factor for display filter (0=max, 1=min goes with par 57)","fbnr":1,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Dynamic display factor","max":1,"min":0,"name":"DynDispFct","parameter":56,"poll":False,"process":117,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0,000001","description":"static display factor for display filter (0=max, 1=min goes with par 56)","fbnr":2,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Static display factor","max":1,"min":0,"name":"StaDispFct","parameter":57,"poll":False,"process":117,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"calibration mode selection (not active until cntrlmode has been set to value 9)","fbnr":1,"group0":7,"group1":"","group2":"","highly secured":False,"longname":"Calibration mode","max":255,"min":0,"name":"CalMode","parameter":58,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"61000","description":"valve offset: amount of DAC steps within 1 potmeter step","fbnr":7,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve offset","max":65535,"min":-32767,"name":"ValveOffst","parameter":59,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"7","description":"monitor: output signal (measure)  selection for bus and analog output","fbnr":2,"group0":2,"group1":"","group2":"","highly secured":False,"longname":"Monitor mode","max":255,"min":0,"name":"Monitor","parameter":60,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"alarm register containing warning flags","fbnr":2,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Alarm register1","max":"","min":"","name":"AlarmReg1","parameter":61,"poll":False,"process":114,"read":True,"secured":True,"varlength":8,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"alarm register containing critical error flags","fbnr":3,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Alarm register2","max":"","min":"","name":"AlarmReg2","parameter":62,"poll":False,"process":114,"read":True,"secured":True,"varlength":8,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"210A7D","description":"calibration register zero scale input 1 ADC","fbnr":1,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"<CalRegZS1>","max":16777215,"min":0,"name":"CalRegZS1","parameter":63,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"52A513","description":"calibration register full scale input 1 ADC","fbnr":2,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"<CalRegFS1>","max":16777215,"min":0,"name":"CalRegFS1","parameter":64,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"210A7D","description":"calibration register zero scale input 2 ADC","fbnr":3,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"<CalRegZS2>","max":16777215,"min":0,"name":"CalRegZS2","parameter":65,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"52A513","description":"calibration register full scale input 2 ADC","fbnr":4,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"<CalRegFS2>","max":16777215,"min":0,"name":"CalRegFS2","parameter":66,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"18904E","description":"ADC control register","fbnr":4,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"ADC control register","max":4294967295,"min":0,"name":"ADCcntrReg","parameter":67,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"sensor bridge zero potmeter setting","fbnr":5,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"Bridge potmeter","max":255,"min":0,"name":"BridgePotm","parameter":68,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"broadcast alarm messsage enable flag","fbnr":4,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"<AlarmEnble>","max":255,"min":0,"name":"AlarmEnble","parameter":69,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"test mode selection  (not active until cntrlmode has been set to value 5)","fbnr":5,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Test mode","max":255,"min":0,"name":"TestMode","parameter":70,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"channel selection ADC","fbnr":6,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"<ADC channel select>","max":32,"min":1,"name":"ChanSelect","parameter":71,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"controller response for normal steps (128=normal, <128=slower, >128=faster)","fbnr":5,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Normal step controller response","max":255,"min":0,"name":"ContrResp","parameter":72,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1.0","description":"analog input filter constant (0=max, 1=min)","fbnr":3,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Setpoint exponential smoothing filter","max":1,"min":0,"name":"ErrInFilCo","parameter":73,"poll":False,"process":117,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1.0","description":"sensor input filter constant (0=max, 1=min)","fbnr":4,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Sensor exponential smoothing filter","max":1,"min":0,"name":"ExpSmooCon","parameter":74,"poll":False,"process":117,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"32767","description":"analog output correction factor zero scale (meas outp DSCM-A 0=0 other 32767=0)","fbnr":21,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog output zero scale","max":65535,"min":-32767,"name":"AnOutCorZS","parameter":75,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"2000","description":"analog output correction factor full scale (meas outp 2000 = 1 * multiplication)","fbnr":22,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog output full scale","max":65535,"min":0,"name":"AnOutCorFS","parameter":76,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"32767","description":"analog input correction factor zero scale (ext setp DSCM-A 0=0 other 32767=0)","fbnr":23,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog input zero scale","max":65535,"min":-32767,"name":"AnInpCorZS","parameter":77,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"2000","description":"analog input correction factor full scale (ext setp 2000 = 1 * multiplication)","fbnr":24,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog input full scale","max":65535,"min":0,"name":"AnInpCorFS","parameter":78,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"(auto)tuning mode selection (not active until cntrlmode has been set to value 6)","fbnr":7,"group0":7,"group1":"","group2":"","highly secured":False,"longname":"Tuning mode","max":255,"min":0,"name":"TuningMode","parameter":79,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"valve type (needed for controlling behaviour)","fbnr":6,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve default","max":255,"min":0,"name":"DefVlvType","parameter":80,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"contains number(s) of changed processes for indirect polling (0xXX / 0xFF)","fbnr":19,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Global modify","max":255,"min":0,"name":"GlobModify","parameter":81,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0.1","description":"correction factor valve curve ratio high/low area","fbnr":7,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve span correction factor","max":1,"min":0,"name":"SpanCorr","parameter":82,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"20,80","description":"Valve curve correction for controller (max. factor*0.1, flow where factor = 1)","fbnr":8,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve curve correction","max":"","min":"","name":"VlvCrvStps","parameter":83,"poll":False,"process":114,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"10,5000,10,5000","description":"array with memberships for normal Fuzzy controller","fbnr":9,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"<MemShipNor>","max":"","min":"","name":"MemShipNor","parameter":84,"poll":False,"process":114,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1,10000,3750,20000","description":"array with memberships for 0-open Fuzzy controller","fbnr":10,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"<MemShipOpn>","max":"","min":"","name":"MemShipOpn","parameter":85,"poll":False,"process":114,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"4","description":"IO status byte for jumper settings and LED signal modes","fbnr":11,"group0":12,"group1":20,"group2":"","highly secured":False,"longname":"IO status","max":255,"min":0,"name":"IOStatus","parameter":86,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"DMFC","description":"(FLOW-BUS) device type information string","fbnr":1,"group0":13,"group1":"","group2":"","highly secured":False,"longname":"Device type","max":"","min":"","name":"DeviceType","parameter":90,"poll":False,"process":113,"read":True,"secured":False,"varlength":6,"vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"F201C-FA","description":"BHT model number  information string","fbnr":2,"group0":13,"group1":"","group2":"","highly secured":False,"longname":"BHT model number","max":"","min":"","name":"ModelNum","parameter":91,"poll":False,"process":113,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"SN999999A","description":"serial number information string","fbnr":3,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Serial number","max":"","min":"","name":"SerialNum","parameter":92,"poll":False,"process":113,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"STANDARD","description":"customer model number information string","fbnr":4,"group0":11,"group1":"","group2":"","highly secured":False,"longname":"Customer model number","max":"","min":"","name":"MfrConfig","parameter":93,"poll":False,"process":113,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"01,01,95","description":"special BHT parameter (to be changed by BHT only)","fbnr":1,"group0":14,"group1":"","group2":"","highly secured":True,"longname":"BHT1","max":"","min":"","name":"BHT1","parameter":94,"poll":False,"process":118,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"special BHT parameter","fbnr":2,"group0":14,"group1":"","group2":"","highly secured":False,"longname":"BHT2","max":65535,"min":0,"name":"BHT2","parameter":95,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"special BHT parameter","fbnr":3,"group0":14,"group1":"","group2":"","highly secured":False,"longname":"BHT3","max":3000000000,"min":-3000000000,"name":"BHT3","parameter":96,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"special BHT parameter","fbnr":4,"group0":14,"group1":"","group2":"","highly secured":False,"longname":"BHT4","max":65535,"min":0,"name":"BHT4","parameter":97,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"special BHT parameter","fbnr":5,"group0":14,"group1":"","group2":"","highly secured":False,"longname":"BHT5","max":255,"min":0,"name":"BHT5","parameter":98,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"special BHT parameter","fbnr":6,"group0":14,"group1":"","group2":"","highly secured":False,"longname":"BHT6","max":255,"min":0,"name":"BHT6","parameter":99,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"special BHT parameter","fbnr":7,"group0":14,"group1":"","group2":"","highly secured":False,"longname":"BHT7","max":255,"min":0,"name":"BHT7","parameter":100,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"special BHT parameter","fbnr":8,"group0":14,"group1":"","group2":"","highly secured":False,"longname":"BHT8","max":255,"min":0,"name":"BHT8","parameter":101,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"special BHT parameter","fbnr":9,"group0":14,"group1":"","group2":"","highly secured":False,"longname":"BHT9","max":3000000000,"min":-3000000000,"name":"BHT9","parameter":102,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"special BHT parameter (to be changed by BHT only)","fbnr":10,"group0":14,"group1":"","group2":"","highly secured":True,"longname":"BHT10","max":1,"min":0,"name":"BHT10","parameter":103,"poll":False,"process":118,"read":False,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"128","description":"broadcast repeating time (x5 ms) (old:height of open at 0 pulse train for valve)","fbnr":16,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Broadcast repeating time","max":255,"min":0,"name":"PulseHight","parameter":104,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"VX.XX","description":"revision number of firmware","fbnr":5,"group0":13,"group1":20,"group2":"","highly secured":False,"longname":"Firmware version","max":"","min":"","name":"Version","parameter":105,"poll":False,"process":113,"read":True,"secured":False,"varlength":6,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"type of pressure sensor","fbnr":9,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Pressure sensor type","max":255,"min":0,"name":"PressSensr","parameter":106,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1013.25","description":"mbar atmospheric (central) barometer pressure","fbnr":8,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Barometer pressure","max":1200,"min":0,"name":"BaroPress","parameter":107,"poll":False,"process":116,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"32767","description":"analog sensor signal input corr. factor zero scale (DSCM-A 0=0 other 32767=0)","fbnr":25,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Sensor input zero scale","max":65535,"min":-32767,"name":"AnIn1CorZS","parameter":108,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"2000","description":"analog sensor signal input correction factor full scale (2000=1*multiplication)","fbnr":26,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Sensor input full scale","max":65535,"min":0,"name":"AnIn1CorFS","parameter":109,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"32767","description":"analog Vref input correction factor zero scale (DSCM-A 0=0 other 32767=0)","fbnr":27,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Reference voltage input zero scale","max":65535,"min":-32767,"name":"AnIn2CorZS","parameter":110,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"2000","description":"analog Vref input correction factor full scale (2000=1*multiplication)","fbnr":28,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Reference voltage input full scale","max":65535,"min":0,"name":"AnIn2CorFS","parameter":111,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"32767","description":"analog setpoint output correction factor zero scale (DSCM-A 0=0 other 32767=0)","fbnr":29,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog setpoint zero scale","max":65535,"min":-32767,"name":"AnOu1CorZS","parameter":112,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"2000","description":"analog setpoint output correction factor full scale (2000=1*multiplication)","fbnr":30,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog setpoint full scale","max":65535,"min":0,"name":"AnOu1CorFS","parameter":113,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"reset facilities (program/alarm/batchcounter)","fbnr":8,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Reset","max":255,"min":0,"name":"Reset","parameter":114,"poll":False,"process":115,"read":False,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"USERTAG","description":"user definable alias string","fbnr":6,"group0":11,"group1":"","group2":"","highly secured":False,"longname":"User tag","max":"","min":"","name":"UserTag","parameter":115,"poll":False,"process":113,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"maximum limit for sensor signal to trigger alarm situation","fbnr":1,"group0":15,"group1":"","group2":"","highly secured":False,"longname":"Alarm limit maximum","max":41600,"min":0,"name":"AlrmMaxLim","parameter":116,"poll":False,"process":97,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"minimum limit for sensor signal to trigger alarm situation","fbnr":2,"group0":15,"group1":"","group2":"","highly secured":False,"longname":"Alarm limit minimum","max":41600,"min":0,"name":"AlrmMinLim","parameter":117,"poll":False,"process":97,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"alarm mode","fbnr":3,"group0":15,"group1":"","group2":"","highly secured":False,"longname":"Alarm mode","max":255,"min":0,"name":"AlrmMode","parameter":118,"poll":False,"process":97,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"alarm relais activity mode during alarm situation","fbnr":4,"group0":15,"group1":"","group2":"","highly secured":False,"longname":"Alarm output mode","max":255,"min":0,"name":"AlrmOutMod","parameter":119,"poll":False,"process":97,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"setpoint change enable during alarm situation","fbnr":5,"group0":15,"group1":"","group2":"","highly secured":False,"longname":"Alarm setpoint mode","max":1,"min":0,"name":"AlrmStpMod","parameter":120,"poll":False,"process":97,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"new/safe setpoint during alarm situation (until reset)","fbnr":6,"group0":15,"group1":"","group2":"","highly secured":False,"longname":"Alarm new setpoint","max":32767,"min":0,"name":"AlrmNwSetp","parameter":121,"poll":False,"process":97,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"actual counter value","fbnr":1,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter value","max":3.40282E38,"min":0,"name":"CntrValue","parameter":122,"poll":True,"process":104,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"pointer to counter unit (classic counter unit table)","fbnr":2,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter unit index","max":31,"min":0,"name":"CntrUnit","parameter":123,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"counter limit/batch","fbnr":3,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter limit","max":3.40282E38,"min":0,"name":"CntrLimit","parameter":124,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"counter relais activity mode when limit/batch has been reached","fbnr":4,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter output mode","max":255,"min":0,"name":"CntrOutMod","parameter":125,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"setpoint change enable during counter limit/batch situation (until reset)","fbnr":5,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter setpoint mode","max":1,"min":0,"name":"CntrStpMod","parameter":126,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"new/safe setpoint at counter limit/batch situation (until reset) (normally = 0%)","fbnr":6,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter new setpoint","max":32767,"min":0,"name":"CntrNwSetp","parameter":127,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"ln","description":"counter readout unit (informative only for older devices)","fbnr":7,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter unit","max":"","min":"","name":"CntrUntstr","parameter":128,"poll":False,"process":104,"read":True,"secured":False,"varlength":4,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"ln/min","description":"capacity readout unit (informative only for older devices)","fbnr":31,"group0":3,"group1":19,"group2":"","highly secured":False,"longname":"Capacity unit","max":"","min":"","name":"capunitstr","parameter":129,"poll":False,"process":"","read":True,"secured":True,"varlength":7,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"counter mode","fbnr":8,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter mode","max":255,"min":0,"name":"CntrMode","parameter":130,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"VX.XX","description":"minimum required hardware revision level for firmware version","fbnr":7,"group0":13,"group1":"","group2":"","highly secured":False,"longname":"Minimum hardware revision","max":"","min":"","name":"HwRev","parameter":131,"poll":False,"process":113,"read":True,"secured":False,"varlength":1,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"1.0","description":"readout factor for direct reading (changes with readunit: local on module, R.O.)","fbnr":1,"group0":17,"group1":"","group2":"","highly secured":False,"longname":"<RCreadfact>","max":10000000000,"min":1E-10,"name":"RCreadfact","parameter":132,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"1","description":"channel number for operation","fbnr":2,"group0":17,"group1":"","group2":"","highly secured":False,"longname":"<channumber>","max":120,"min":1,"name":"channumber","parameter":133,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"master channel for master-slave operation","fbnr":3,"group0":17,"group1":"","group2":"","highly secured":False,"longname":"<masterchan>","max":120,"min":0,"name":"masterchan","parameter":134,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"32000","description":"RC slave factor","fbnr":4,"group0":17,"group1":"","group2":"","highly secured":False,"longname":"<RCslavefct>","max":32000,"min":0,"name":"RCslavefct","parameter":135,"poll":False,"process":"","read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"3","description":"physical node address for channel number","fbnr":5,"group0":17,"group1":"","group2":"","highly secured":False,"longname":"<inputnode>","max":128,"min":0,"name":"inputnode","parameter":136,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"physical process for channel number","fbnr":6,"group0":17,"group1":"","group2":"","highly secured":False,"longname":"<inputproc>","max":128,"min":0,"name":"inputproc","parameter":137,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"ln/min","description":"readout unit for direct reading (local variable on module: read only)","fbnr":7,"group0":17,"group1":"","group2":"","highly secured":False,"longname":"<RCreadunit>","max":"","min":"","name":"RCreadunit","parameter":138,"poll":False,"process":"","read":True,"secured":False,"varlength":7,"vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"100.0","description":"slave factor for master slave control (setp = master output * slave factor)","fbnr":1,"group0":18,"group1":"","group2":"","highly secured":False,"longname":"Slave factor","max":500,"min":0,"name":"SlaveFact%","parameter":139,"poll":False,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"reference voltage input for setpoint signal","fbnr":2,"group0":18,"group1":"","group2":"","highly secured":False,"longname":"Reference voltage input","max":65535,"min":0,"name":"VrefInput","parameter":140,"poll":True,"process":33,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"controller response when controller is stable: |measure-setpoint| < 2%","fbnr":17,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Stable situation controller response","max":255,"min":0,"name":"RespStable","parameter":141,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"20","description":"absolute temperature in degrees Celsius","fbnr":7,"group0":19,"group1":13,"group2":"","highly secured":False,"longname":"Temperature","max":500,"min":-250,"name":"temperatur","parameter":142,"poll":True,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1013.25","description":"absolute pressure in mbar (FPP, FRM, FTM) or bar","fbnr":8,"group0":19,"group1":13,"group2":"","highly secured":False,"longname":"Pressure","max":3.40282E38,"min":-3.40282E38,"name":"pressure","parameter":143,"poll":True,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"time in milliseconds","fbnr":9,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Time","max":3.40282E38,"min":0,"name":"time","parameter":144,"poll":True,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"50","description":"calibrated volume in litres","fbnr":10,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Calibrated volume","max":3.40282E38,"min":0,"name":"calvolume","parameter":145,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"pointer to sensor number in calibration tube (FPP)","fbnr":16,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Sensor number","max":4,"min":0,"name":"sensornr","parameter":146,"poll":True,"process":"","read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Piston Prover operation mode (write) and status information (read back)","fbnr":10,"group0":20,"group1":19,"group2":"","highly secured":False,"longname":"Range select","max":99,"min":0,"name":"rangeselct","parameter":147,"poll":True,"process":115,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"maximum admitted duration time for specific procedure (in 100 ms)","fbnr":2,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Time out","max":30000,"min":0,"name":"TimeOut","parameter":148,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"frequency in Hz","fbnr":9,"group0":21,"group1":"","group2":"","highly secured":False,"longname":"Frequency","max":100000,"min":0,"name":"frequency","parameter":149,"poll":True,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"42773.4","description":"For FRM and FTM imp/m3 and for FCM imp/kg","fbnr":10,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Impulses/m3","max":3.40282E38,"min":0,"name":"imp/m3","parameter":150,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"volume flow referenced to normal conditions i.e. 0 �C, 1013.25 hPa(a) in ln/min","fbnr":5,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Normal volume flow","max":3.40282E38,"min":0,"name":"RefVolFlow","parameter":151,"poll":False,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"volume flow at actual conditions in l/min","fbnr":6,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Volume flow","max":3.40282E38,"min":-3.40282E38,"name":"volumeflow","parameter":152,"poll":False,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"relative pressure between atmosphere and sensor position","fbnr":11,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Delta-p","max":100000,"min":-100000,"name":"delta-p","parameter":153,"poll":True,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"scaling factor (multiplication) for readout on display (for optimal resolution)","fbnr":13,"group0":21,"group1":"","group2":"","highly secured":False,"longname":"<scalefact>","max":10000,"min":1,"name":"scalefact","parameter":154,"poll":False,"process":33,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"SENSOR0","description":"label with information about stop sensor","fbnr":17,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Sensor name","max":"","min":"","name":"sensorname","parameter":155,"poll":False,"process":"","read":True,"secured":True,"varlength":10,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"15","description":"enable reset of alarm by: keyboard, external signal, FLOW-BUS, automatic","fbnr":9,"group0":15,"group1":"","group2":"","highly secured":False,"longname":"Reset alarm enable","max":255,"min":0,"name":"RstAlarmEn","parameter":156,"poll":False,"process":97,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"7","description":"enable reset of counter by: keyboard, external signal, FLOW-BUS, automatic","fbnr":9,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Reset counter enable","max":255,"min":0,"name":"RstCountEn","parameter":157,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"3","description":"node number of master instrument output signal for a slave","fbnr":14,"group0":18,"group1":"","group2":"","highly secured":False,"longname":"Master node","max":128,"min":1,"name":"MasterNode","parameter":158,"poll":False,"process":33,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1","description":"process number of master instrument output signal for a slave","fbnr":15,"group0":18,"group1":"","group2":"","highly secured":False,"longname":"Master process","max":128,"min":1,"name":"MasterProc","parameter":159,"poll":False,"process":33,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"3","description":"node number of instrument to be operated by another module (keyboard/display)","fbnr":16,"group0":18,"group1":"","group2":"","highly secured":False,"longname":"Remote instrument node","max":128,"min":1,"name":"InstrNode","parameter":160,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"process number of instrument to be operated by another module (keyboard/display)","fbnr":17,"group0":18,"group1":"","group2":"","highly secured":False,"longname":"Remote instrument process","max":128,"min":1,"name":"InstrProc","parameter":161,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0.0","description":"Mnimum value at 0% for special user readout unit","fbnr":18,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Minimum custom range","max":3.40282E38,"min":-3.40282E38,"name":"RangeMin","parameter":162,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"100.0","description":"Maximum value at 100% for special user readout unit","fbnr":20,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Maximum custom range","max":3.40282E38,"min":-3.40282E38,"name":"RangeMax","parameter":163,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Relay/TTL output setting (disabled when used by alarm or counter)","fbnr":11,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Relay/TTL output","max":255,"min":0,"name":"Relay/TTL","parameter":164,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Controller response when valve opens from zero","fbnr":18,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Open from zero controller response","max":255,"min":0,"name":"RespOpen0%","parameter":165,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Controller settings for special purpose","fbnr":20,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Controller features","max":255,"min":0,"name":"ContrType","parameter":166,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"10","description":"PID factor Kp","fbnr":21,"group0":8,"group1":3,"group2":"","highly secured":False,"longname":"PID-Kp","max":3.40282E38,"min":0,"name":"PIDKp","parameter":167,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0.05","description":"PID factor Ti","fbnr":22,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"PID-Ti","max":3.40282E38,"min":0,"name":"PIDTi","parameter":168,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"PID factor Td","fbnr":23,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"PID-Td","max":3.40282E38,"min":0,"name":"PIDTd","parameter":169,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1.293","description":"Density of selected fluid in kg/m3","fbnr":21,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Density","max":3.40282E38,"min":0,"name":"Density","parameter":170,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"","description":"Number of calibration certificate (last basic calibration)","fbnr":8,"group0":13,"group1":3,"group2":"","highly secured":True,"longname":"Calibration certificate","max":"","min":"","name":"CalCertNr","parameter":171,"poll":False,"process":113,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"19991231","description":"Date of last (basic) calibration","fbnr":9,"group0":13,"group1":3,"group2":"","highly secured":True,"longname":"Calibration date","max":"","min":"","name":"CalDate","parameter":172,"poll":False,"process":113,"read":True,"secured":True,"varlength":8,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"000000","description":"Service number for repair/rebuilding/recalibration","fbnr":10,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Service number","max":"","min":"","name":"ServiceNr","parameter":173,"poll":False,"process":113,"read":True,"secured":True,"varlength":15,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"19991231","description":"Date of last service action","fbnr":11,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Service date","max":"","min":"","name":"ServDate","parameter":174,"poll":False,"process":113,"read":True,"secured":True,"varlength":8,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"7","description":"Identification number (type) of instrument/device","fbnr":12,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Identification number","max":255,"min":0,"name":"IdentNr","parameter":175,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"special BHT parameter (to be changed by BHT only)","fbnr":11,"group0":14,"group1":"","group2":"","highly secured":True,"longname":"BHT11","max":255,"min":0,"name":"BHT11","parameter":176,"poll":False,"process":118,"read":False,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"power supply indication in Vdc","fbnr":12,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Power mode","max":50,"min":0,"name":"PowerMode","parameter":177,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"3","description":"pressure inlet (upstream) of fluid in bara (for first fluidnr only)","fbnr":13,"group0":13,"group1":3,"group2":"","highly secured":False,"longname":"Pressure inlet","max":100000,"min":-100000,"name":"Pupstream","parameter":178,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"pressure outlet (downstream) of fluid in bara (for first fluidnr only)","fbnr":14,"group0":13,"group1":3,"group2":"","highly secured":False,"longname":"Pressure outlet","max":100000,"min":-100000,"name":"Pdownstrm","parameter":179,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"orifice diameter in mm","fbnr":15,"group0":13,"group1":"","group2":"","highly secured":False,"longname":"Orifice","max":1000,"min":0,"name":"Orifice","parameter":180,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"20","description":"temperature of fluid through instrument (for first fluidnr only)","fbnr":16,"group0":13,"group1":3,"group2":"","highly secured":False,"longname":"Fluid temperature","max":3.40282E38,"min":-273.15,"name":"FluidTemp","parameter":181,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"3","description":"time alarm and reset action will be delayed when alarm limit has been exceeded","fbnr":7,"group0":15,"group1":"","group2":"","highly secured":False,"longname":"Alarm delay","max":255,"min":0,"name":"AlrmDelay","parameter":182,"poll":False,"process":97,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"capacity of instrument at zero 0% in sensor base units (mostly equal to zero)","fbnr":22,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Capacity 0%","max":3.40282E38,"min":-3.40282E38,"name":"capacity0%","parameter":183,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"number of instrument channels available for this device","fbnr":18,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Number of channels","max":120,"min":0,"name":"NumOfChan","parameter":184,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"5","description":"function of device","fbnr":20,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Device function","max":255,"min":0,"name":"DeviceFunc","parameter":185,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"1","description":"Channel  number to scan with real time information (to be set once)","fbnr":1,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Scan channel","max":255,"min":1,"name":"ScanChan","parameter":186,"poll":False,"process":123,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"8","description":"Parameter number to scan with real time information (to be set once)","fbnr":3,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Scan parameter","max":255,"min":0,"name":"ScanPar","parameter":187,"poll":False,"process":123,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"50","description":"Scan interval time in msec between two samples (to be set once)","fbnr":4,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Scan time","max":65535,"min":0,"name":"ScanTime","parameter":188,"poll":False,"process":123,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"Scanned data with time label (can be readout event by event)","fbnr":10,"group0":22,"group1":4,"group2":"","highly secured":False,"longname":"Scan data","max":"","min":"","name":"ScanData","parameter":189,"poll":False,"process":123,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0.04","description":"First-step offset current/voltage for valve when opening from 0%","fbnr":24,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve open","max":24,"min":0,"name":"ValveOpen","parameter":190,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1","description":"Amount of runs of a piston prover (0 = stability check)","fbnr":13,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Number of runs","max":255,"min":0,"name":"NrOfRuns","parameter":191,"poll":False,"process":115,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"10","description":"Minimum process time of a piston prover in 0.1 seconds","fbnr":14,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Minimum process time","max":255,"min":0,"name":"MinProTime","parameter":192,"poll":False,"process":115,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0.0001","description":"Leak rate piston prover/piston tare value multiplier","fbnr":9,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Leak rate","max":3,"min":0,"name":"LeakRate","parameter":193,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"Sets instr. in info mode for 1 read-cycle to check available parameter options","fbnr":15,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Mode info request","max":"","min":"","name":"ModeInfReq","parameter":194,"poll":False,"process":115,"read":True,"secured":True,"varlength":4,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"Gives info about possible values of a mode in an array as result of ModeInfo req","fbnr":16,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Mode info option list","max":"","min":"","name":"ModeInfOpt","parameter":195,"poll":False,"process":115,"read":True,"secured":False,"varlength":255,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"","description":"Gives description about one of the mode options","fbnr":17,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Mode info option description","max":"","min":"","name":"ModeInfDes","parameter":196,"poll":False,"process":115,"read":True,"secured":False,"varlength":255,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Enables/disables options for calibration device (8 bits for 8 options)","fbnr":18,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Calibrations options","max":255,"min":0,"name":"CalType","parameter":197,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Real mass flow in kg/min","fbnr":4,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Mass flow","max":3.40282E38,"min":-3.40282E38,"name":"MassFlow","parameter":198,"poll":True,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"2","description":"Fieldbus1 address (main interface)","fbnr":10,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 address","max":255,"min":0,"name":"BusAddress","parameter":199,"poll":False,"process":125,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Configuration setting for interface to other bus-systems","fbnr":3,"group0":23,"group1":"","group2":"","highly secured":True,"longname":"Interface configuration","max":3,"min":0,"name":"InterfConf","parameter":200,"poll":False,"process":125,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"12000000","description":"Fieldbus1 baudrate (main interface)","fbnr":9,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 baudrate","max":10000000000,"min":0,"name":"Baudrate","parameter":201,"poll":False,"process":125,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"Fieldbus1 diagnose string (main interface)","fbnr":20,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 diagnostic string","max":"","min":"","name":"BusDiagnos","parameter":202,"poll":False,"process":125,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"10","description":"Number of vanes for use in a rotor meter","fbnr":22,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Number of vanes","max":255,"min":0,"name":"NrOfVanes","parameter":203,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"FLOW-BUS","description":"Fieldbus1 name (main interface)","fbnr":21,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 name","max":"","min":"","name":"Fieldbus","parameter":204,"poll":False,"process":125,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"measured value for direct reading (in capunits, max.= capacity)","fbnr":0,"group0":2,"group1":"","group2":"","highly secured":False,"longname":"fMeasure","max":3.40282E38,"min":-3.40282E38,"name":"fMeasure","parameter":205,"poll":True,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"setpoint: wanted value for direct reading (in capunits, max.= capacity)","fbnr":3,"group0":2,"group1":18,"group2":"","highly secured":False,"longname":"fSetpoint","max":3.40282E38,"min":-3.40282E38,"name":"fSetpoint","parameter":206,"poll":False,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Mass in g","fbnr":23,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Mass","max":3.40282E38,"min":-3.40282E38,"name":"Mass","parameter":207,"poll":True,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"","description":"Manufacturer Status register (64 diagnostic bits)","fbnr":1,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Manufacturer status register","max":"","min":"","name":"Mstatus","parameter":208,"poll":False,"process":119,"read":True,"secured":False,"varlength":8,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"","description":"Manufacturer Warning register (64 diagnostic bits)","fbnr":2,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Manufacturer warning register","max":"","min":"","name":"Mwarning","parameter":209,"poll":False,"process":119,"read":True,"secured":False,"varlength":8,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"","description":"Manufacturer Error register (64 diagnostic bits)","fbnr":3,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Manufacturer error register","max":"","min":"","name":"Merror","parameter":210,"poll":False,"process":119,"read":True,"secured":False,"varlength":8,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"","description":"Diagnostic history string (contains history of diag codes)","fbnr":4,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Diagnostic history string","max":"","min":"","name":"DiagHist","parameter":211,"poll":False,"process":119,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Diagnostic mode (0 = diagnostics off, 1 = diagnostics on)","fbnr":5,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Diagnostic mode","max":255,"min":0,"name":"DiagMode","parameter":212,"poll":False,"process":119,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Manufacturer Status enable (0-127 or 254 = disable all, 255 = enable all)","fbnr":6,"group0":4,"group1":"","group2":"","highly secured":False,"longname":"Manufacturer status enable","max":255,"min":0,"name":"MStatEnabl","parameter":213,"poll":False,"process":119,"read":False,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Analog measure output, zero adjust","fbnr":21,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog output zero adjust","max":3.40282E38,"min":-3.40282E38,"name":"AnOutZA","parameter":214,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Analog measure output, span adjust","fbnr":22,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog output span adjust","max":3.40282E38,"min":-3.40282E38,"name":"AnOutSA","parameter":215,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Analog setpoint input, zero adjust","fbnr":23,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog input zero adjust","max":3.40282E38,"min":-3.40282E38,"name":"AnInZA","parameter":216,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Analog setpoint input, span adjust","fbnr":24,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Analog input span adjust","max":3.40282E38,"min":-3.40282E38,"name":"AnInSA","parameter":217,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Sensor input, zero adjust","fbnr":25,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Sensor input zero adjust","max":3.40282E38,"min":-3.40282E38,"name":"SensInZA","parameter":218,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Sensor input, span adjust","fbnr":26,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Sensor input span adjust","max":3.40282E38,"min":-3.40282E38,"name":"SensInSA","parameter":219,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Sensor temperature input, zero adjust","fbnr":27,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Temperature input zero adjust","max":3.40282E38,"min":-3.40282E38,"name":"TempInZA","parameter":220,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Sensor temperature input, span adjust","fbnr":28,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Temperature input span adjust","max":3.40282E38,"min":-3.40282E38,"name":"TempInSA","parameter":221,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1.0","description":"Sensor input filter adapt setting","fbnr":5,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Adaptive smoothing factor","max":1,"min":0,"name":"ExpSmooAd","parameter":222,"poll":False,"process":117,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"32000","description":"Slope setpoint step. Setpoint step for the given slopetime","fbnr":24,"group0":18,"group1":"","group2":"","highly secured":False,"longname":"Slope setpoint step","max":32000,"min":0,"name":"SlopeSetp","parameter":223,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Number of samples for average filter","fbnr":6,"group0":6,"group1":"","group2":"","highly secured":False,"longname":"Filter length","max":255,"min":0,"name":"FilterLen","parameter":224,"poll":False,"process":117,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Actual accuracy in current unit","fbnr":25,"group0":2,"group1":"","group2":"","highly secured":False,"longname":"Absolute accuracy","max":3.40282E38,"min":-3.40282E38,"name":"fAccuracy","parameter":225,"poll":False,"process":33,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Lookup table for linearisation index (x and y direction)","fbnr":26,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Lookup table index","max":255,"min":0,"name":"LookI","parameter":226,"poll":False,"process":33,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Lookup table for linearisation x","fbnr":27,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Lookup table X","max":3.40282E38,"min":-3.40282E38,"name":"LookX","parameter":227,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Lookup table for linearisation y","fbnr":28,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Lookup table Y","max":3.40282E38,"min":-3.40282E38,"name":"LookY","parameter":228,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Lookup table for linearisation at certain temperature index (z direction)","fbnr":29,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Lookup table temperature index","max":255,"min":0,"name":"LookTempI","parameter":229,"poll":False,"process":33,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Lookup table for linearisation at certain temperature (z)","fbnr":30,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Lookup table temperature","max":3.40282E38,"min":-273.15,"name":"LookTemp","parameter":230,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0.2","description":"Maximum current/voltage for valve","fbnr":25,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve maximum","max":24,"min":0,"name":"ValveMax","parameter":231,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Valve output mode selection","fbnr":26,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve mode","max":255,"min":0,"name":"ValveMode","parameter":232,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0.96","description":"Valve open current/voltage correction (example: 0.96, Open = ValveOpen * 0.96)","fbnr":27,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve open correction","max":1.5,"min":0,"name":"VlvOpenCor","parameter":233,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Valve hold current/voltage at %0 setp (example: 0.8, Hold = ValveOpen * 0.8)","fbnr":28,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve zero hold","max":1,"min":0,"name":"VlvZeroHld","parameter":234,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0.009","description":"Valve slope time (seconds)","fbnr":29,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve slope","max":50,"min":0,"name":"ValveSlope","parameter":235,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"IFI data dump protocol communication string","fbnr":21,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"IFI data","max":"","min":"","name":"IFIData","parameter":236,"poll":False,"process":0,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Piston Prover information about used sensors","fbnr":20,"group0":20,"group1":19,"group2":"","highly secured":False,"longname":"Range used","max":99,"min":0,"name":"RangeUsed","parameter":237,"poll":True,"process":115,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Fluidset properties","fbnr":31,"group0":3,"group1":"","group2":"","highly secured":True,"longname":"Fluidset properties","max":255,"min":0,"name":"FldSetProp","parameter":238,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Lookup table unit type","fbnr":12,"group0":3,"group1":"","group2":"","highly secured":True,"longname":"Lookup table unit type index","max":255,"min":0,"name":"LUnitType","parameter":239,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"Lookup table unit type name","fbnr":13,"group0":3,"group1":"","group2":"","highly secured":True,"longname":"Lookup table unit type","max":"","min":"","name":"LUnTypName","parameter":240,"poll":False,"process":33,"read":True,"secured":True,"varlength":20,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Lookup table unit (unit LUTy)","fbnr":16,"group0":3,"group1":"","group2":"","highly secured":True,"longname":"Lookup table unit index","max":255,"min":0,"name":"LUnit","parameter":241,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"kg/s","description":"Lookup table unit name","fbnr":17,"group0":3,"group1":"","group2":"","highly secured":True,"longname":"Lookup table unit","max":"","min":"","name":"LUnitName","parameter":242,"poll":False,"process":33,"read":True,"secured":True,"varlength":7,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Capacity (readout) unit type","fbnr":29,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Capacity unit type index","max":255,"min":0,"name":"CUnitType","parameter":243,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"","description":"Capacity (readout) unit type name","fbnr":30,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Capacity unit type","max":"","min":"","name":"CUnTypName","parameter":244,"poll":False,"process":"","read":True,"secured":True,"varlength":20,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Capacity (readout) unit type temperature (�C)","fbnr":10,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Capacity unit type temperature","max":3.40282E38,"min":-273.15,"name":"CUnTypTemp","parameter":245,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1","description":"Capacity (readout) unit type pressure (bar (a))","fbnr":11,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Capacity unit type pressure","max":3.40282E38,"min":0,"name":"CUnTypPres","parameter":246,"poll":False,"process":33,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Minimum capacity in output capacity units","fbnr":27,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Capacity minimum","max":3.40282E38,"min":-3.40282E38,"name":"CapMin","parameter":247,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1","description":"Maximum capacity in output capacity units","fbnr":28,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Capacity maximum","max":3.40282E38,"min":-3.40282E38,"name":"CapMax","parameter":248,"poll":False,"process":"","read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Formula type needed for conversion","fbnr":17,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Formula type","max":65535,"min":0,"name":"FormulaTyp","parameter":249,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Heat capacity (Cp) (sensor conditions)","fbnr":18,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Heat capacity","max":3.40282E38,"min":0,"name":"HeatCap","parameter":250,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Thermal conductivity (sensor conditions)","fbnr":20,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Thermal conductivity","max":3.40282E38,"min":0,"name":"ThermCond","parameter":251,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Dynamic viscosity (fluid conditions)","fbnr":21,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Viscosity","max":3.40282E38,"min":0,"name":"Viscosity","parameter":252,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Standard mass flow in ln/min (20�C, 1.01325 bar (a)) air or g/h H2O equivalent","fbnr":22,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Standard flow","max":3.40282E38,"min":-3.40282E38,"name":"NormMasFlw","parameter":253,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1","description":"Controller speed factor (gain)","fbnr":30,"group0":8,"group1":3,"group2":"","highly secured":False,"longname":"Controller speed","max":3.40282E38,"min":0,"name":"Kspeed","parameter":254,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Sensor code","fbnr":23,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Sensor code","max":65535,"min":0,"name":"SensorCode","parameter":255,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Sensor configuration code","fbnr":24,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Sensor configuration code","max":255,"min":0,"name":"SensorRevC","parameter":256,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Restriction code","fbnr":25,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Restriction code","max":65535,"min":0,"name":"RestrCode","parameter":257,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Restriction configuration code","fbnr":26,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Restriction configurator code","max":255,"min":0,"name":"RestrRevC","parameter":258,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Restriction NxP (proportional to air equivalent capacity of LFE)","fbnr":27,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Restriction NxP","max":2147483648,"min":0,"name":"RestrNxP","parameter":259,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"V,V","description":"Seals information (1st byte = other, 2nd = plunger seal)","fbnr":28,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Seals information","max":255,"min":0,"name":"Seals","parameter":260,"poll":False,"process":113,"read":True,"secured":True,"varlength":16,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Valve code","fbnr":29,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Valve code","max":65535,"min":0,"name":"ValveCode","parameter":261,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Valve configuration code","fbnr":30,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Valve configuration code","max":255,"min":0,"name":"ValveRevC","parameter":262,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Instrument properties","fbnr":31,"group0":13,"group1":"","group2":"","highly secured":True,"longname":"Instrument properties","max":2147483648,"min":0,"name":"InstrProp","parameter":263,"poll":False,"process":113,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Lookup table for frequency index","fbnr":10,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Lookup table frequency index","max":1,"min":0,"name":"LookFreqI","parameter":264,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Lookup table for frequency frequency","fbnr":11,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Lookup table frequency frequency","max":3.40282E38,"min":-3.40282E38,"name":"LFFreq","parameter":265,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Lookup table for frequency temperature","fbnr":12,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Lookup table frequency temperature","max":3.40282E38,"min":-273.15,"name":"LFTemp","parameter":266,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Lookup table for frequency density","fbnr":13,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Lookup table frequency density","max":3.40282E38,"min":-3.40282E38,"name":"LFDensity","parameter":267,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Lookup table for frequency span adjust","fbnr":14,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Lookup table frequency span adjust","max":3.40282E38,"min":-3.40282E38,"name":"LFSpanAdj","parameter":268,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Capacity (readout) unit index (extended unit table)","fbnr":15,"group0":3,"group1":19,"group2":"","highly secured":False,"longname":"Capacity unit index (ext)","max":255,"min":0,"name":"CUnit","parameter":269,"poll":False,"process":65,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Actual density, measured by instrument","fbnr":15,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Density actual","max":3.40282E38,"min":-3.40282E38,"name":"DensityAct","parameter":270,"poll":True,"process":116,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Measured restriction","fbnr":18,"group0":13,"group1":"","group2":"","highly secured":False,"longname":"Measured restriction","max":3.40282E38,"min":0,"name":"RestrMeas","parameter":271,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Potmeter for sensor temperature compensation","fbnr":8,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Temperature potmeter","max":255,"min":0,"name":"TempPotm","parameter":272,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Gain for sensor temperature compensation","fbnr":9,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Temperature potmeter gain","max":255,"min":0,"name":"TempGain","parameter":273,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1","description":"Counter controller overrun correction","fbnr":10,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter controller overrun correction","max":3.40282E38,"min":0,"name":"CntrCConv","parameter":274,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"1","description":"Counter controller gain","fbnr":11,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter controller gain","max":3.40282E38,"min":0,"name":"CntrCGain","parameter":275,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Sub fluid number","fbnr":1,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Sub fluid number","max":255,"min":0,"name":"FluidSub","parameter":276,"poll":False,"process":65,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Temperature compensation factor","fbnr":17,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"Temperature compensation factor","max":3.40282E38,"min":-3.40282E38,"name":"FreqTCor","parameter":277,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"DSP register address","fbnr":29,"group0":13,"group1":"","group2":"","highly secured":False,"longname":"DSP register address","max":4294967295,"min":0,"name":"DSPRegI","parameter":278,"poll":False,"process":116,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"c","write":True},
{"advanced":True,"available":True,"default":"0","description":"DSP register long","fbnr":30,"group0":13,"group1":"","group2":"","highly secured":False,"longname":"DSP register long","max":4294967295,"min":-4294967295,"name":"DSPRegLng","parameter":279,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"DSP register floating point","fbnr":30,"group0":13,"group1":"","group2":"","highly secured":False,"longname":"DSP register floating point","max":3.40282E38,"min":-3.40282E38,"name":"DSPRegFlt","parameter":280,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"DSP register integer","fbnr":31,"group0":13,"group1":"","group2":"","highly secured":False,"longname":"DSP register integer","max":65535,"min":-65535,"name":"DSPRegInt","parameter":281,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Standard deviation","fbnr":0,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Standard deviation","max":3.40282E38,"min":-3.40282E38,"name":"StdDev","parameter":282,"poll":True,"process":121,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Measurement status","fbnr":1,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Measurement status","max":65535,"min":0,"name":"MeasStatus","parameter":283,"poll":False,"process":121,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Measurement stop criteria","fbnr":2,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Measurement stop criteria","max":65535,"min":0,"name":"MStopCrit","parameter":284,"poll":False,"process":121,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Measurement time out","fbnr":3,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Measurement time out","max":65535,"min":0,"name":"MTimeOut","parameter":285,"poll":False,"process":121,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Measurement maximum number of runs","fbnr":4,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Maximum number of runs","max":65535,"min":0,"name":"MMaxNrRuns","parameter":286,"poll":False,"process":121,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Measurement minimum standard deviation","fbnr":5,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Minimum standard deviation","max":3.40282E38,"min":-3.40282E38,"name":"MMinStdDev","parameter":287,"poll":False,"process":121,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"IO status for switches","fbnr":31,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"IO switch status","max":4294967295,"min":0,"name":"IOSwitchSt","parameter":288,"poll":False,"process":114,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"513","description":"Sensor bridge settings","fbnr":21,"group0":9,"group1":"","group2":"","highly secured":True,"longname":"Sensor bridge settings","max":65535,"min":0,"name":"SensBridge","parameter":295,"poll":False,"process":65,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Sensor bridge current (0..0.02A)","fbnr":22,"group0":9,"group1":"","group2":"","highly secured":True,"longname":"Sensor bridge current","max":3.40282E38,"min":0,"name":"SensCurren","parameter":296,"poll":False,"process":65,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Sensor resistance (Ohm)","fbnr":23,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"Sensor resistance","max":3.40282E38,"min":0,"name":"SensResist","parameter":297,"poll":False,"process":65,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Sensor bridge voltage (0..11.5V)","fbnr":24,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"Sensor bridge voltage","max":3.40282E38,"min":0,"name":"SensVolt","parameter":298,"poll":False,"process":65,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"","description":"Sensor name (based on sensor detection / sensor resistance)","fbnr":25,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"Sensor group name","max":"","min":"","name":"SensName","parameter":299,"poll":False,"process":65,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Sensor temperature at calibration","fbnr":20,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"Sensor calibration temperature","max":3.40282E38,"min":-3.40282E38,"name":"SensCalTmp","parameter":300,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Valve safe state","fbnr":31,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve safe state","max":255,"min":0,"name":"ValveSafe","parameter":301,"poll":False,"process":115,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Counter unit type","fbnr":12,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter unit type index","max":255,"min":0,"name":"CnUnitType","parameter":302,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"","description":"Counter unit type name","fbnr":13,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter unit type","max":"","min":"","name":"CnUnTypNam","parameter":303,"poll":False,"process":104,"read":True,"secured":False,"varlength":20,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Counter unit index (extended counter unit table)","fbnr":14,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter unit index (ext)","max":255,"min":0,"name":"CnUnit","parameter":304,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"255","description":"Fieldbus1 select (main interface)","fbnr":8,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 selection","max":255,"min":0,"name":"Bus1Select","parameter":305,"poll":False,"process":125,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Fieldbus1 medium id (main interface)","fbnr":11,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 medium","max":255,"min":0,"name":"Bus1Medium","parameter":306,"poll":False,"process":125,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Fieldbus2 mode","fbnr":7,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus2 mode","max":255,"min":0,"name":"Bus2Mode","parameter":307,"poll":False,"process":124,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"2","description":"Fieldbus2 select","fbnr":8,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus2 selection","max":255,"min":0,"name":"Bus2Select","parameter":308,"poll":False,"process":124,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"2","description":"Fieldbus2 address","fbnr":10,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus2 address","max":255,"min":0,"name":"Bus2Addr","parameter":309,"poll":False,"process":124,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"12000000","description":"Fieldbus2 baudrate","fbnr":9,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus2 baudrate","max":10000000000,"min":0,"name":"Bus2Baud","parameter":310,"poll":False,"process":124,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Fieldbus2 medium id","fbnr":11,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus2 medium","max":255,"min":0,"name":"Bus2Medium","parameter":311,"poll":False,"process":124,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"Fieldbus2 diagnose string","fbnr":20,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus2 diagnostics","max":"","min":"","name":"Bus2Diagn","parameter":312,"poll":False,"process":124,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"Propar","description":"Fieldbus2 name","fbnr":21,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus2 name","max":"","min":"","name":"Bus2Name","parameter":313,"poll":False,"process":124,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"PIO channel selection","fbnr":0,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO channel selection","max":255,"min":0,"name":"PIOChSel","parameter":314,"poll":False,"process":120,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO parameter connected to i/o channel (=process number*256+parameter number)","fbnr":2,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO parameter","max":65535,"min":0,"name":"PIOParm","parameter":315,"poll":False,"process":120,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO input/output filter constant (0..1, 0=max, 1=min)","fbnr":6,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO input/output filter","max":3.40282E38,"min":-3.40282E38,"name":"PIOFilter","parameter":316,"poll":False,"process":120,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO parameter capacity, 0% value","fbnr":7,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO parameter capacity 0%","max":3.40282E38,"min":-3.40282E38,"name":"PIOPrmCap0","parameter":317,"poll":False,"process":120,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO parameter capacity, 100% value","fbnr":3,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO parameter capacity 100%","max":3.40282E38,"min":-3.40282E38,"name":"PIOPrmCap","parameter":318,"poll":False,"process":120,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO i/o channel configuration selection","fbnr":1,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO configuration selection","max":255,"min":0,"name":"PIOCfgSel","parameter":319,"poll":False,"process":120,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO analog input/output, zero adjust value","fbnr":4,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO analog zero adjust","max":3.40282E38,"min":-3.40282E38,"name":"PIOAnZA","parameter":320,"poll":False,"process":120,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO analog input/output, span adjust factor","fbnr":5,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO analog span adjust","max":3.40282E38,"min":-3.40282E38,"name":"PIOAnSA","parameter":321,"poll":False,"process":120,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO max hardware capacity (max capacity value in i/o units, V, A, Hz, etc.)","fbnr":8,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO hardware capacity max","max":3.40282E38,"min":-3.40282E38,"name":"PIOHwCapMx","parameter":322,"poll":False,"process":120,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO capacity set selection","fbnr":9,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO capacity set selection","max":255,"min":0,"name":"PIOCapSel","parameter":323,"poll":False,"process":120,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO hardware capacity, 0% value (capacity value in i/o units, V, A, Hz, etc.)","fbnr":10,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO hardware capacity 0%","max":3.40282E38,"min":-3.40282E38,"name":"PIOHwCap0","parameter":324,"poll":False,"process":120,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"PIO hardware capacity, 100% value (capacity value in i/o units, V, A, Hz, etc.)","fbnr":11,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"PIO hardware capacity 100%","max":3.40282E38,"min":-3.40282E38,"name":"PIOHwCap","parameter":325,"poll":False,"process":120,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Hardware platform identification number","fbnr":6,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Hardware platform id","max":65535,"min":0,"name":"HardwId","parameter":326,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Hardware platform sub identification number","fbnr":7,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Hardware platform sub id","max":255,"min":0,"name":"HardwSubId","parameter":327,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Temporary volatile fieldbus2 baudrate","fbnr":31,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Temporary baudrate","max":10000000000,"min":0,"name":"TempBaud","parameter":328,"poll":False,"process":124,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Monitor mode for setpoint","fbnr":23,"group0":2,"group1":"","group2":"","highly secured":False,"longname":"Setpoint monitor mode","max":255,"min":0,"name":"SetMonitor","parameter":329,"poll":False,"process":115,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special BHT parameter","fbnr":12,"group0":14,"group1":"","group2":"","highly secured":False,"longname":"BHT12","max":10000000000,"min":0,"name":"BHT12","parameter":330,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Nominal sensor voltage (used for sensor temperature voltage compensation)","fbnr":26,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"Nominal sensor voltage","max":3.40282E38,"min":0,"name":"SenNomVolt","parameter":331,"poll":False,"process":65,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Sensor voltage compensation factor (used for sensor temperature voltage comp.)","fbnr":16,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"Sensor voltage compensation factor","max":3.40282E38,"min":-3.40282E38,"name":"VoltTCor","parameter":332,"poll":False,"process":116,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"PCB serial number (unique number or string written by PCB manufacturer)","fbnr":31,"group0":4,"group1":"","group2":"","highly secured":True,"longname":"PCB serial number","max":"","min":"","name":"PCBSerial","parameter":333,"poll":False,"process":119,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Minimum measure time between sensors of a piston prover in 0.1 seconds","fbnr":24,"group0":20,"group1":"","group2":"","highly secured":False,"longname":"Minimum measure time","max":255,"min":0,"name":"MinMTime","parameter":334,"poll":False,"process":115,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Fieldbus1 parity (main interface)","fbnr":12,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 parity","max":255,"min":0,"name":"Bus1Parity","parameter":335,"poll":False,"process":125,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Fieldbus2 parity","fbnr":12,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus2 parity","max":255,"min":0,"name":"Bus2Parity","parameter":336,"poll":False,"process":124,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Firmware identification number","fbnr":8,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Firmware id","max":255,"min":0,"name":"FirmwId","parameter":337,"poll":False,"process":0,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Valve 100% current/voltage","fbnr":13,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Valve 100%","max":3.40282E38,"min":0,"name":"Valve100","parameter":338,"poll":False,"process":114,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Minimum setpoint for controller","fbnr":14,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Setpoint minimum","max":1,"min":0,"name":"SetpMin","parameter":339,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"PCB serial number index","fbnr":30,"group0":4,"group1":0,"group2":0,"highly secured":False,"longname":"PCB serial number index","max":255,"min":0,"name":"PCBSerialI","parameter":340,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"00-00-00-00-00-00","description":"Fieldbus1 MAC address","fbnr":13,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 MAC address","max":"","min":"","name":"Bus1MAC","parameter":341,"poll":False,"process":125,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Index of source fluidset for conversion","fbnr":0,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Source fluidset index","max":255,"min":0,"name":"FluidNrSrc","parameter":342,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Calculation method for capacity range","fbnr":1,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Capacity range calculation method","max":65535,"min":0,"name":"CapRanCalM","parameter":343,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Actual fluid property index","fbnr":2,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Actual fluid property index","max":255,"min":0,"name":"ActFlPIdx","parameter":344,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Actual fluid property calculation method","fbnr":3,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Actual fluid property calculation method","max":65535,"min":0,"name":"ActFlPCalM","parameter":345,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Mix fraction type","fbnr":4,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Mix fraction type","max":255,"min":0,"name":"MixFrcType","parameter":346,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Mix volume type temperature (�C)","fbnr":5,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Mix volume temperature","max":3.40282E38,"min":-273.15,"name":"MixVolTemp","parameter":347,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Mix volume type pressure (bar (a))","fbnr":6,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Mix volume pressure","max":3.40282E38,"min":0,"name":"MixVolPres","parameter":348,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Mix component index","fbnr":7,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Mix component index","max":255,"min":0,"name":"MixCompIdx","parameter":349,"poll":False,"process":126,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Mix component fraction","fbnr":8,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Mix component fraction","max":1,"min":0,"name":"MixCompFrc","parameter":350,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"AIR","description":"Mix component fluid name","fbnr":9,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Mix component fluid name","max":"","min":"","name":"MixCompNam","parameter":351,"poll":False,"process":126,"read":True,"secured":True,"varlength":10,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"AIR","description":"Fluid name for equivalent measure","fbnr":10,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Equivalent measure fluid name","max":"","min":"","name":"EqMFldName","parameter":352,"poll":False,"process":126,"read":True,"secured":False,"varlength":10,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Equivalent measure at actual conditions (kg/s)","fbnr":11,"group0":19,"group1":"","group2":"","highly secured":False,"longname":"Equivalent measure","max":3.40282E38,"min":-3.40282E38,"name":"EqMeasure","parameter":353,"poll":False,"process":126,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Fluid conversion conditions (temperature and pressure) selection","fbnr":12,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Fluid conversion conditions selection","max":65535,"min":0,"name":"FlConCondS","parameter":354,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Sensor+restriction capacity minimum","fbnr":13,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Sensor+restriction capacity minimum","max":3.40282E38,"min":-3.40282E38,"name":"SensCapMin","parameter":355,"poll":False,"process":126,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Sensor+restriction capacity maximum","fbnr":14,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Sensor+restriction capacity maximum","max":3.40282E38,"min":-3.40282E38,"name":"SensCapMax","parameter":356,"poll":False,"process":126,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Valve capacity minimum","fbnr":15,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Valve capacity minimum","max":3.40282E38,"min":-3.40282E38,"name":"ValvCapMin","parameter":357,"poll":False,"process":126,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Valve capacity maximum","fbnr":16,"group0":3,"group1":"","group2":"","highly secured":False,"longname":"Valve capacity maximum","max":3.40282E38,"min":-3.40282E38,"name":"ValvCapMax","parameter":358,"poll":False,"process":126,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Maximum allowed dosing time (s)","fbnr":15,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Maximum allowed dosing time","max":3.40282E38,"min":0,"name":"MaxDosTime","parameter":359,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Most recent dosing time (s)","fbnr":16,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Most recent dosing time","max":3.40282E38,"min":0,"name":"DosingTime","parameter":360,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Controller hysteresis","fbnr":15,"group0":8,"group1":"","group2":"","highly secured":False,"longname":"Controller hysteresis","max":1,"min":0,"name":"CtrlHyst","parameter":361,"poll":False,"process":114,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Pressure sensor input, zero adjust","fbnr":17,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Pressure sensor input zero adjust","max":3.40282E38,"min":-3.40282E38,"name":"PressInZA","parameter":362,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"1","description":"Pressure sensor input, span adjust","fbnr":18,"group0":10,"group1":"","group2":"","highly secured":False,"longname":"Pressure sensor input span adjust","max":3.40282E38,"min":-3.40282E38,"name":"PressInSA","parameter":363,"poll":False,"process":126,"read":True,"secured":True,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter index","fbnr":0,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter index","max":2147483648,"min":0,"name":"SpecialIdx","parameter":364,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter i long","fbnr":1,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter i long","max":2147483648,"min":-2147483648,"name":"SpecLong","parameter":365,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter i float","fbnr":2,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter i float","max":3.40282E38,"min":-3.40282E38,"name":"SpecFloat","parameter":366,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"Special parameter i name","fbnr":3,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter i name","max":"","min":"","name":"SpecName","parameter":367,"poll":False,"process":127,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter l1 long","fbnr":4,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter l1 long","max":2147483648,"min":-2147483648,"name":"SpecLong1","parameter":368,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter l2 long","fbnr":5,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter l2 long","max":2147483648,"min":-2147483648,"name":"SpecLong2","parameter":369,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter l3 long","fbnr":6,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter l3 long","max":2147483648,"min":-2147483648,"name":"SpecLong3","parameter":370,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter l4 long","fbnr":7,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter l4 long","max":2147483648,"min":-2147483648,"name":"SpecLong4","parameter":371,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter l5 long","fbnr":8,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter l5 long","max":2147483648,"min":-2147483648,"name":"SpecLong5","parameter":372,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter f1 float","fbnr":9,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter f1 float","max":3.40282E38,"min":-3.40282E38,"name":"SpecFloat1","parameter":373,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter f2 float","fbnr":10,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter f2 float","max":3.40282E38,"min":-3.40282E38,"name":"SpecFloat2","parameter":374,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter f3 float","fbnr":11,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter f3 float","max":3.40282E38,"min":-3.40282E38,"name":"SpecFloat3","parameter":375,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter f4 float","fbnr":12,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter f4 float","max":3.40282E38,"min":-3.40282E38,"name":"SpecFloat4","parameter":376,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Special parameter f5 float","fbnr":13,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Special parameter f5 float","max":3.40282E38,"min":-3.40282E38,"name":"SpecFloat5","parameter":377,"poll":False,"process":127,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Fieldbus interface index","fbnr":7,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus interface index","max":255,"min":0,"name":"BusIntIdx","parameter":378,"poll":False,"process":125,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Selftest status","fbnr":7,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest status","max":2147483648,"min":0,"name":"SelfStat","parameter":379,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Selftest block index","fbnr":8,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest block index","max":255,"min":0,"name":"SelfBIdx","parameter":380,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Selftest test index","fbnr":9,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest test index","max":65535,"min":0,"name":"SelfTIdx","parameter":381,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Selftest test description/name","fbnr":10,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest test description/name","max":"","min":"","name":"SelfName","parameter":382,"poll":False,"process":119,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Selftest test result (pass/fail)","fbnr":11,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest test result","max":255,"min":0,"name":"SelfRes","parameter":383,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Selftest test lower limit","fbnr":12,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest test lower limit","max":3.40282E38,"min":-3.40282E38,"name":"SelfLower","parameter":384,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Selftest test upper limit","fbnr":13,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest test upper limit","max":3.40282E38,"min":-3.40282E38,"name":"SelfUpper","parameter":385,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Selftest test lowest  value","fbnr":14,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest test lowest  value","max":3.40282E38,"min":-3.40282E38,"name":"SelfLow","parameter":386,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Selftest test highest  value","fbnr":15,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest test highest  value","max":3.40282E38,"min":-3.40282E38,"name":"SelfHigh","parameter":387,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Selftest test average value","fbnr":16,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest test average value","max":3.40282E38,"min":-3.40282E38,"name":"SelfAvg","parameter":388,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Selftest test standard deviation","fbnr":17,"group0":12,"group1":"","group2":"","highly secured":False,"longname":"Selftest test standard deviation","max":3.40282E38,"min":-3.40282E38,"name":"SelfStDv","parameter":389,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0.0.0.0","description":"IP address, when no address configured value is 0.0.0.0 (DHCP)","fbnr":14,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 IP address","max":"","min":"","name":"Bus1IP","parameter":390,"poll":False,"process":125,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0.0.0.0","description":"Subnet mask for the IP address","fbnr":15,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 subnet mask","max":"","min":"","name":"Bus1SubMas","parameter":391,"poll":False,"process":125,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0.0.0.0","description":"IP address of the gateway","fbnr":16,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 gateway address","max":"","min":"","name":"Bus1GateW","parameter":392,"poll":False,"process":125,"read":True,"secured":True,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Totalizer value","fbnr":17,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Totalizer value","max":3.4E40,"min":-3.4E40,"name":"TotalzrVal","parameter":393,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"ln/min","description":"Totalizer unit","fbnr":18,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Totalizer unit","max":"","min":"","name":"TotalzrUnt","parameter":394,"poll":False,"process":104,"read":True,"secured":True,"varlength":4,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Counter threshold","fbnr":20,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Counter threshold","max":3.4E40,"min":-3.4E40,"name":"CntrThresh","parameter":395,"poll":False,"process":104,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"","description":"Contains fieldbus firmware version information","fbnr":22,"group0":23,"group1":"","group2":"","highly secured":False,"longname":"Fieldbus1 version","max":"","min":"","name":"Bus1Ver","parameter":396,"poll":False,"process":125,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Relative humidity (0..100%)","fbnr":20,"group0":9,"group1":"","group2":"","highly secured":False,"longname":"Relative humidity","max":100,"min":0,"name":"RelHum","parameter":397,"poll":False,"process":65,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Dosing type","fbnr":1,"group0":16,"group1":"","group2":"","highly secured":False,"longname":"Dosing type","max":255,"min":0,"name":"DosType","parameter":398,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Dosing controller type","fbnr":2,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Dosing controller type","max":255,"min":0,"name":"DosCtrType","parameter":399,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Dosing actuator type","fbnr":3,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Dosing actuator type","max":255,"min":0,"name":"DosActType","parameter":400,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Dosing start type","fbnr":4,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Dosing start type","max":255,"min":0,"name":"DosStartTp","parameter":401,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Batch dosing start delay time","fbnr":5,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Batch dosing start delay time","max":3.4E40,"min":0,"name":"BtchDelayT","parameter":402,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Batch dosing time","fbnr":6,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Batch dosing time","max":3.4E40,"min":0,"name":"BtchDosT","parameter":403,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Batch dosing repetition interval","fbnr":7,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Batch dosing repetition interval","max":3.4E40,"min":0,"name":"BtchDosRep","parameter":404,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Batch size","fbnr":8,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Batch size","max":3.4E40,"min":0,"name":"BtchSize","parameter":405,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Batch accuracy [%]","fbnr":9,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Batch accuracy [%]","max":3.4E40,"min":0,"name":"BtchAcc","parameter":406,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Actual batch size","fbnr":10,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Actual batch size","max":3.4E40,"min":0,"name":"ActBtchSze","parameter":407,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Actual batch dosing time [s]","fbnr":11,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Actual batch dosing time [s]","max":3.4E40,"min":0,"name":"ActBtchDos","parameter":408,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Actual batch accuracy [%]","fbnr":12,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Actual batch accuracy [%]","max":3.4E40,"min":0,"name":"ActBtchAcc","parameter":409,"poll":False,"process":105,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Batch counter unit","fbnr":21,"group0":16,"group1":0,"group2":0,"highly secured":False,"longname":"Batch counter unit","max":"","min":"","name":"BtchCntUnt","parameter":410,"poll":False,"process":104,"read":True,"secured":False,"varlength":4,"vartype":"c","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Diagnostic newest event index","fbnr":14,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Diagnostic newest event index","max":65535,"min":0,"name":"DiaNEvtIdx","parameter":411,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Diagnostic event index","fbnr":15,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Diagnostic event index","max":65535,"min":0,"name":"DiaEvtIdx","parameter":412,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Diagnostic event code","fbnr":16,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Diagnostic event code","max":65535,"min":0,"name":"DiaEvtCode","parameter":413,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Diagnostic event description","fbnr":20,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Diagnostic event description","max":"","min":"","name":"DiaEvtDesc","parameter":414,"poll":False,"process":118,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Diagnostic event active","fbnr":17,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Diagnostic event active","max":255,"min":0,"name":"DiaEvtAct","parameter":415,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Diagnostic event namur status","fbnr":18,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Diagnostic event namur status","max":255,"min":0,"name":"DiaEvtNSt","parameter":416,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Diagnostic event timestamp","fbnr":21,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Diagnostic event timestamp","max":4294967295,"min":0,"name":"DiaEvtTime","parameter":417,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"l","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Instrument namur status","fbnr":0,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Instrument namur status","max":255,"min":0,"name":"InstrNSt","parameter":418,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Instrument timestamp","fbnr":13,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Instrument timestamp","max":4294967295,"min":0,"name":"InstrTime","parameter":419,"poll":False,"process":118,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Operational history parameter index","fbnr":26,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Operational history parameter index","max":65535,"min":0,"name":"OpHisIdx","parameter":420,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"i","vartype2":"","write":True},
{"advanced":False,"available":True,"default":"0","description":"Operational history parameter name","fbnr":27,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Operational history parameter name","max":"","min":"","name":"OpHisName","parameter":421,"poll":False,"process":118,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Operational history parameter minimum value","fbnr":28,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Operational history parameter minimum value","max":3.4E40,"min":0,"name":"OpHisMin","parameter":422,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Operational history parameter maximum value","fbnr":29,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Operational history parameter maximum value","max":3.4E40,"min":0,"name":"OpHisMax","parameter":423,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Operational history parameter average","fbnr":30,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Operational history parameter average","max":3.4E40,"min":0,"name":"OpHisAvg","parameter":424,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Operational history parameter standard deviation","fbnr":31,"group0":24,"group1":0,"group2":0,"highly secured":False,"longname":"Operational history parameter standard deviation","max":3.4E40,"min":0,"name":"OpHisStdDv","parameter":425,"poll":False,"process":118,"read":True,"secured":False,"varlength":"","vartype":"f","vartype2":"","write":False},
{"advanced":True,"available":True,"default":"0","description":"Fieldbus stop bits","fbnr":23,"group0":23,"group1":0,"group2":0,"highly secured":False,"longname":"Fieldbus stop bits","max":255,"min":0,"name":"Bus1StopB","parameter":426,"poll":False,"process":125,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Fieldbus connection mode","fbnr":24,"group0":23,"group1":0,"group2":0,"highly secured":False,"longname":"Fieldbus connection mode","max":255,"min":0,"name":"Bus1ConMod","parameter":427,"poll":False,"process":125,"read":True,"secured":True,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Fieldbus passkey","fbnr":25,"group0":23,"group1":0,"group2":0,"highly secured":False,"longname":"Fieldbus passkey","max":4294967295,"min":0,"name":"Bus1PssKey","parameter":428,"poll":False,"process":125,"read":True,"secured":True,"varlength":"","vartype":"l","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"","description":"Feature unlock code","fbnr":27,"group0":12,"group1":0,"group2":0,"highly secured":False,"longname":"Feature unlock code","max":"","min":"","name":"FeatUnlckC","parameter":429,"poll":False,"process":119,"read":True,"secured":False,"varlength":-2,"vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Feature selection","fbnr":28,"group0":12,"group1":0,"group2":0,"highly secured":False,"longname":"Feature selection","max":255,"min":0,"name":"FeatSel","parameter":430,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True},
{"advanced":True,"available":True,"default":"0","description":"Feature state","fbnr":29,"group0":12,"group1":0,"group2":0,"highly secured":False,"longname":"Feature state","max":255,"min":0,"name":"FeatState","parameter":431,"poll":False,"process":119,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":False},
{"advanced":False,"available":True,"default":"0","description":"Control function","fbnr":10,"group0":8,"group1":0,"group2":0,"highly secured":False,"longname":"Control function","max":255,"min":0,"name":"CtrlFunc","parameter":432,"poll":False,"process":115,"read":True,"secured":False,"varlength":"","vartype":"c","vartype2":"","write":True}], 
"parvalue":[{"description":"temporary busmaster","filter":"","id":6,"name":"arbitrage","parameter":6,"value":1},
{"description":"setpoint = BUS setpoint","filter":"","id":12,"name":"cntrlmode","parameter":12,"value":0},
{"description":"pressure (controller)","filter":"","id":22,"name":"sensortype","parameter":22,"value":0},
{"description":"no error message in alarm error status register","filter":"&H01","id":28,"name":"alarminfo","parameter":28,"value":0},
{"description":"OFF","filter":"","id":44,"name":"opermodeTA","parameter":44,"value":0},
{"description":"0...5 Vdc operation","filter":"&H3F","id":53,"name":"AnalogMode","parameter":53,"value":0},
{"description":"idle: no action","filter":"","id":58,"name":"CalMode","parameter":58,"value":0},
{"description":"(filtered) setpoint","filter":"","id":60,"name":"monitor","parameter":60,"value":0},
{"description":"disable","filter":"","id":69,"name":"AlarmEnble","parameter":69,"value":0},
{"description":"idle; no action","filter":"","id":70,"name":"TestMode","parameter":70,"value":0},
{"description":"AD channel 1","filter":"","id":71,"name":"ChanSelect","parameter":71,"value":1},
{"description":"idle; no action","filter":"","id":79,"name":"TuningMode","parameter":79,"value":0},
{"description":"normally closed","filter":"","id":80,"name":"DefVlvType","parameter":80,"value":0},
{"description":"delta-P 0..5\" W.C.","filter":"","id":106,"name":"PressSensr","parameter":106,"value":0},
{"description":"no reset","filter":"","id":114,"name":"Reset","parameter":114,"value":0},
{"description":"off","filter":"","id":118,"name":"AlrmMode","parameter":118,"value":0},
{"description":"no relais activity at alarm","filter":"","id":119,"name":"AlrmOutMod","parameter":119,"value":0},
{"description":"no setpoint change at alarm","filter":"","id":120,"name":"AlrmStpMod","parameter":120,"value":0},
{"description":"no relais activity at batch limit","filter":"","id":125,"name":"CntrOutMod","parameter":125,"value":0},
{"description":"setpoint change at batch limit disabled","filter":"","id":126,"name":"CntrStpMod","parameter":126,"value":0},
{"description":"off","filter":"","id":130,"name":"CntrMode","parameter":130,"value":0},
{"description":"setpoint = analog input","filter":"","id":140,"name":"cntrlmode","parameter":12,"value":1},
{"description":"setpoint =  master output(FLOW-BUS) * slave factor(FLOW-BUS)","filter":"","id":141,"name":"cntrlmode","parameter":12,"value":2},
{"description":"close valve","filter":"","id":142,"name":"cntrlmode","parameter":12,"value":3},
{"description":"controller idle (no reaction on changes in sensor signal)","filter":"","id":143,"name":"cntrlmode","parameter":12,"value":4},
{"description":"testmode enable (select subject with par 70)","filter":"","id":144,"name":"cntrlmode","parameter":12,"value":5},
{"description":"tuningmode enable (select subtject with par 79)","filter":"","id":145,"name":"cntrlmode","parameter":12,"value":6},
{"description":"setpoint = 100%","filter":"","id":146,"name":"cntrlmode","parameter":12,"value":7},
{"description":"purge valve (fully open)","filter":"","id":147,"name":"cntrlmode","parameter":12,"value":8},
{"description":"calibration mode enable (select subject with par 58)","filter":"","id":148,"name":"cntrlmode","parameter":12,"value":9},
{"description":"setpoint = master output(analog in) * slave factor(FLOW-BUS)","filter":"","id":149,"name":"cntrlmode","parameter":12,"value":10},
{"description":"setpoint = keyboard OR FLOW-BUS setpoint","filter":"","id":150,"name":"cntrlmode","parameter":12,"value":11},
{"description":"setpoint  = 0%","filter":"","id":151,"name":"cntrlmode","parameter":12,"value":12},
{"description":"liquid volume (controller)","filter":"","id":152,"name":"sensortype","parameter":22,"value":1},
{"description":"liquid/gas mass (controller)","filter":"","id":153,"name":"sensortype","parameter":22,"value":2},
{"description":"gas volume (controller)","filter":"","id":154,"name":"sensortype","parameter":22,"value":3},
{"description":"other sensor type (controller)","filter":"","id":156,"name":"sensortype","parameter":22,"value":4},
{"description":"pressure (sensor)","filter":"","id":157,"name":"sensortype","parameter":22,"value":128},
{"description":"liquid volume (sensor)","filter":"","id":158,"name":"sensortype","parameter":22,"value":129},
{"description":"liquid/gas mass (sensor)","filter":"","id":159,"name":"sensortype","parameter":22,"value":130},
{"description":"gas volume (sensor)","filter":"","id":160,"name":"sensortype","parameter":22,"value":131},
{"description":"other sensor type (sensor)","filter":"","id":161,"name":"sensortype","parameter":22,"value":132},
{"description":"analog output = 0 %","filter":"","id":162,"name":"CalMode","parameter":58,"value":15},
{"description":"analog output = 100 %","filter":"","id":163,"name":"CalMode","parameter":58,"value":16},
{"description":"normal sensor signal (Output)","filter":"","id":164,"name":"monitor","parameter":60,"value":8},
{"description":"analog input signal","filter":"","id":165,"name":"monitor","parameter":60,"value":9},
{"description":"power supply voltage","filter":"","id":166,"name":"monitor","parameter":60,"value":10},
{"description":"controller error input signal / raw sensor signal","filter":"","id":167,"name":"monitor","parameter":60,"value":1},
{"description":"controller output signal to valve","filter":"","id":168,"name":"monitor","parameter":60,"value":2},
{"description":"sensor signal slow","filter":"","id":169,"name":"monitor","parameter":60,"value":3},
{"description":"sensor signal slow filtered","filter":"","id":170,"name":"monitor","parameter":60,"value":4},
{"description":"linearization output","filter":"","id":171,"name":"monitor","parameter":60,"value":5},
{"description":"differentiator output","filter":"","id":172,"name":"monitor","parameter":60,"value":6},
{"description":"differentiator output filtered","filter":"","id":173,"name":"monitor","parameter":60,"value":7},
{"description":"reset counter value (no mode change) or common reset","filter":"","id":174,"name":"Reset","parameter":114,"value":1},
{"description":"reset alarm","filter":"","id":175,"name":"Reset","parameter":114,"value":2},
{"description":"reset counter value (counter off)","filter":"","id":176,"name":"Reset","parameter":114,"value":4},
{"description":"restart batch counter","filter":"","id":177,"name":"Reset","parameter":114,"value":3},
{"description":"alarm on absolute limits","filter":"","id":178,"name":"AlrmMode","parameter":118,"value":1},
{"description":"alarm on limits related to setpoint (response alarm)","filter":"","id":179,"name":"AlrmMode","parameter":118,"value":2},
{"description":"relais pulses until reset","filter":"","id":180,"name":"AlrmOutMod","parameter":119,"value":1},
{"description":"relais activated until reset","filter":"","id":181,"name":"AlrmOutMod","parameter":119,"value":2},
{"description":"new/safe setpoint at alarm enabled (set at par 121)","filter":"","id":182,"name":"AlrmStpMod","parameter":120,"value":1},
{"description":"relais pulses after reaching batch limit until reset","filter":"","id":183,"name":"CntrOutMod","parameter":125,"value":1},
{"description":"relais activated after reaching batch limit until reset","filter":"","id":184,"name":"CntrOutMod","parameter":125,"value":2},
{"description":"counting upwards continuously","filter":"","id":185,"name":"CntrMode","parameter":130,"value":1},
{"description":"counting up to limit (batchcounter)","filter":"","id":186,"name":"CntrMode","parameter":130,"value":2},
{"description":"PC(ISA) interface","filter":"","id":187,"name":"IdentNr","parameter":175,"value":2},
{"description":"ADDA4 (4 channels)","filter":"","id":188,"name":"IdentNr","parameter":175,"value":3},
{"description":"R/C-module, 32 channels","filter":"","id":189,"name":"IdentNr","parameter":175,"value":4},
{"description":"T/A-module","filter":"","id":190,"name":"IdentNr","parameter":175,"value":5},
{"description":"ADDA1: 1 channel ADDA converter module","filter":"","id":191,"name":"IdentNr","parameter":175,"value":6},
{"description":"DMFC: digital mass flow controller","filter":"","id":192,"name":"IdentNr","parameter":175,"value":7},
{"description":"DMFM: digital mass flow meter","filter":"","id":193,"name":"IdentNr","parameter":175,"value":8},
{"description":"DEPC: digital electronic pressure controller","filter":"","id":194,"name":"IdentNr","parameter":175,"value":9},
{"description":"ACT: single actuator","filter":"","id":195,"name":"IdentNr","parameter":175,"value":11},
{"description":"DLFC: digital liquid flow controller","filter":"","id":196,"name":"IdentNr","parameter":175,"value":12},
{"description":"DLFM: digital liquid flow meter","filter":"","id":197,"name":"IdentNr","parameter":175,"value":13},
{"description":"DSCM-A: digital single channel module for analog instruments","filter":"","id":198,"name":"IdentNr","parameter":175,"value":14},
{"description":"DSCM-D: digital single channel module for digital instr.","filter":"","id":199,"name":"IdentNr","parameter":175,"value":15},
{"description":"FRM: FLOW-BUS rotor meter (calibration-instrument)","filter":"","id":200,"name":"IdentNr","parameter":175,"value":16},
{"description":"FTM: FLOW-BUS turbine meter (calibration-instrument)","filter":"","id":201,"name":"IdentNr","parameter":175,"value":17},
{"description":"FPP: FLOW-BUS piston prover/tube (calibration-instrument)","filter":"","id":202,"name":"IdentNr","parameter":175,"value":18},
{"description":"DEPM: digital electronic pressure meter","filter":"","id":203,"name":"IdentNr","parameter":175,"value":10},
{"description":"F/A-module: special version of T/A-module","filter":"","id":204,"name":"IdentNr","parameter":175,"value":19},
{"description":"A: MAX & RESP AUTO; T: UP TO LIMIT","filter":"","id":205,"name":"opermodeTA","parameter":44,"value":1},
{"description":"A: MIN & RESP AUTO; T: UP AND REPEAT","filter":"","id":206,"name":"opermodeTA","parameter":44,"value":2},
{"description":"A: MAX & RESP; T: DOWN FROM LIMIT","filter":"","id":207,"name":"opermodeTA","parameter":44,"value":3},
{"description":"A: MIN & RESP; T: DOWN AND REPEAT","filter":"","id":208,"name":"opermodeTA","parameter":44,"value":4},
{"description":"A: MAXIMUM ALARM; T: ALWAYS UP","filter":"","id":209,"name":"opermodeTA","parameter":44,"value":5},
{"description":"A: MINIMUM ALARM","filter":"","id":210,"name":"opermodeTA","parameter":44,"value":6},
{"description":"A: RESPONSE ALARM","filter":"","id":211,"name":"opermodeTA","parameter":44,"value":7},
{"description":"0...10 Vdc operation","filter":"&H3F","id":218,"name":"AnalogMode","parameter":53,"value":1},
{"description":"0...20 mA operation","filter":"&H3F","id":219,"name":"AnalogMode","parameter":53,"value":2},
{"description":"4...20 mA operation","filter":"&H3F","id":220,"name":"AnalogMode","parameter":53,"value":3},
{"description":"15...20 mA operation","filter":"&H3F","id":221,"name":"AnalogMode","parameter":53,"value":4},
{"description":"adc self calibration","filter":"","id":222,"name":"CalMode","parameter":58,"value":1},
{"description":"dmfc","filter":"","id":223,"name":"CalMode","parameter":58,"value":2},
{"description":"dmfc","filter":"","id":224,"name":"CalMode","parameter":58,"value":3},
{"description":"dmfc","filter":"","id":225,"name":"CalMode","parameter":58,"value":4},
{"description":"dmfc","filter":"","id":226,"name":"CalMode","parameter":58,"value":5},
{"description":"dmfc","filter":"","id":227,"name":"CalMode","parameter":58,"value":6},
{"description":"dmfc","filter":"","id":228,"name":"CalMode","parameter":58,"value":7},
{"description":"dmfc","filter":"","id":229,"name":"CalMode","parameter":58,"value":8},
{"description":"customer zero","filter":"","id":230,"name":"CalMode","parameter":58,"value":9},
{"description":"adjust Vref output by connecting it to analog in","filter":"","id":231,"name":"CalMode","parameter":58,"value":10},
{"description":"adjust analog out by connecting it to analog in","filter":"","id":232,"name":"CalMode","parameter":58,"value":11},
{"description":"adjust valveoutput by connecting it to analog in","filter":"","id":233,"name":"CalMode","parameter":58,"value":12},
{"description":"dmfc","filter":"","id":234,"name":"CalMode","parameter":58,"value":13},
{"description":"dmfc","filter":"","id":235,"name":"CalMode","parameter":58,"value":14},
{"description":"uProcessor","filter":"","id":236,"name":"TestMode","parameter":70,"value":1},
{"description":"IO","filter":"","id":237,"name":"TestMode","parameter":70,"value":2},
{"description":"RAM","filter":"","id":238,"name":"TestMode","parameter":70,"value":3},
{"description":"FRAM","filter":"","id":239,"name":"TestMode","parameter":70,"value":4},
{"description":"ADC","filter":"","id":240,"name":"TestMode","parameter":70,"value":5},
{"description":"DAC","filter":"","id":241,"name":"TestMode","parameter":70,"value":6},
{"description":"sensor","filter":"","id":242,"name":"TestMode","parameter":70,"value":7},
{"description":"valve drive circuit","filter":"","id":243,"name":"TestMode","parameter":70,"value":8},
{"description":"Vref","filter":"","id":244,"name":"TestMode","parameter":70,"value":9},
{"description":"FLOW-BUS","filter":"","id":245,"name":"TestMode","parameter":70,"value":10},
{"description":"calibration","filter":"","id":246,"name":"TestMode","parameter":70,"value":11},
{"description":"keyboard","filter":"","id":247,"name":"TestMode","parameter":70,"value":12},
{"description":"sensor","filter":"","id":248,"name":"TuningMode","parameter":79,"value":1},
{"description":"valve","filter":"","id":249,"name":"TuningMode","parameter":79,"value":2},
{"description":"Fuzzy controller normal operation","filter":"","id":250,"name":"TuningMode","parameter":79,"value":3},
{"description":"Fuzzy controller open at zero","filter":"","id":251,"name":"TuningMode","parameter":79,"value":4},
{"description":"normally opened","filter":"","id":252,"name":"DefVlvType","parameter":80,"value":1},
{"description":"remain position","filter":"","id":253,"name":"DefVlvType","parameter":80,"value":4},
{"description":"read diagnostic jumper (no diagnostics, read/write)","filter":"&H01","id":254,"name":"IOStatus","parameter":86,"value":1},
{"description":"not used","filter":"&H02","id":255,"name":"IOStatus","parameter":86,"value":1},
{"description":"read micro switch (read/write)","filter":"&H08","id":257,"name":"IOStatus","parameter":86,"value":1},
{"description":"diagnostic jumper set (read only)","filter":"&H10","id":260,"name":"IOStatus","parameter":86,"value":1},
{"description":"initialization jumper set (read only)","filter":"&H20","id":261,"name":"IOStatus","parameter":86,"value":1},
{"description":"analog jumper set (read only)","filter":"&H40","id":262,"name":"IOStatus","parameter":86,"value":1},
{"description":"micro switch pressed (read only)","filter":"&H80","id":263,"name":"IOStatus","parameter":86,"value":1},
{"description":"delta-P 0...10\" W.C.","filter":"","id":264,"name":"PressSensr","parameter":106,"value":1},
{"description":"absolute pressure 800-1200 mbar","filter":"","id":265,"name":"PressSensr","parameter":106,"value":2},
{"description":"normally closed inverse controlled","filter":"","id":266,"name":"DefVlvType","parameter":80,"value":2},
{"description":"normally opened inverse controlled","filter":"","id":267,"name":"DefVlvType","parameter":80,"value":3},
{"description":"analog output = 50 %","filter":"","id":269,"name":"CalMode","parameter":58,"value":17},
{"description":"absolute pressure 800-1100 mbar","filter":"","id":270,"name":"PressSensr","parameter":106,"value":3},
{"description":"mass flow in display unit (normally ln/min)","filter":"","id":271,"name":"monitor","parameter":60,"value":11},
{"description":"volume flow in l/min","filter":"","id":272,"name":"monitor","parameter":60,"value":12},
{"description":"temperature in �C","filter":"","id":273,"name":"monitor","parameter":60,"value":13},
{"description":"pressure absolute in mbara","filter":"","id":274,"name":"monitor","parameter":60,"value":14},
{"description":"time in msec/frequency in Hz.","filter":"","id":275,"name":"monitor","parameter":60,"value":15},
{"description":"calibrated volume at actual sensor in ml","filter":"","id":276,"name":"monitor","parameter":60,"value":16},
{"description":"delta-P pressure in mbarg","filter":"","id":277,"name":"monitor","parameter":60,"value":17},
{"description":"atmospheric (barometer) pressure in mbara","filter":"","id":278,"name":"monitor","parameter":60,"value":18},
{"description":"calibration ready/stop","filter":"","id":279,"name":"rangeselct","parameter":147,"value":0},
{"description":"run calibration until stopsensor 1/select range 1","filter":"","id":280,"name":"rangeselct","parameter":147,"value":1},
{"description":"run calibration until stopsensor 2/select range 2","filter":"","id":281,"name":"rangeselct","parameter":147,"value":2},
{"description":"run calibration until stopsensor 3/select range 3","filter":"","id":282,"name":"rangeselct","parameter":147,"value":3},
{"description":"run calibration until stopsensor 4/select range 4","filter":"","id":283,"name":"rangeselct","parameter":147,"value":4},
{"description":"run calibration and select range 5","filter":"","id":284,"name":"rangeselct","parameter":147,"value":5},
{"description":"run calibration with automatic range selection","filter":"","id":285,"name":"rangeselct","parameter":147,"value":9},
{"description":"run until stopsensor 1 until 3 values between limit","filter":"","id":286,"name":"rangeselct","parameter":147,"value":19},
{"description":"run until stopsensor 2 until 3 values between limit","filter":"","id":287,"name":"rangeselct","parameter":147,"value":29},
{"description":"run until stopsensor 3 until 3 values between limit","filter":"","id":288,"name":"rangeselct","parameter":147,"value":39},
{"description":"run until stopsensor 4 until 3 values between limit","filter":"","id":289,"name":"rangeselct","parameter":147,"value":49},
{"description":"run and select range 5 until 3 values between limit","filter":"","id":290,"name":"rangeselct","parameter":147,"value":59},
{"description":"run with auto-select + 3 values between limit","filter":"","id":291,"name":"rangeselct","parameter":147,"value":99},
{"description":"no reset possible","filter":"","id":300,"name":"RstAlarmEn","parameter":156,"value":0},
{"description":"reset: keyboard","filter":"","id":301,"name":"RstAlarmEn","parameter":156,"value":1},
{"description":"reset: external","filter":"","id":302,"name":"RstAlarmEn","parameter":156,"value":2},
{"description":"reset: keyboard or external","filter":"","id":303,"name":"RstAlarmEn","parameter":156,"value":3},
{"description":"reset: FLOW-BUS","filter":"","id":304,"name":"RstAlarmEn","parameter":156,"value":4},
{"description":"reset: FLOW-BUS or keyboard","filter":"","id":305,"name":"RstAlarmEn","parameter":156,"value":5},
{"description":"reset: FLOW-BUS or external","filter":"","id":306,"name":"RstAlarmEn","parameter":156,"value":6},
{"description":"reset: FLOW-BUS or keyboard or external","filter":"","id":307,"name":"RstAlarmEn","parameter":156,"value":7},
{"description":"reset: automatic","filter":"","id":308,"name":"RstAlarmEn","parameter":156,"value":8},
{"description":"reset: automatic or keyboard","filter":"","id":309,"name":"RstAlarmEn","parameter":156,"value":9},
{"description":"reset: automatic or external","filter":"","id":310,"name":"RstAlarmEn","parameter":156,"value":10},
{"description":"reset: automatic or keyboard or external","filter":"","id":311,"name":"RstAlarmEn","parameter":156,"value":11},
{"description":"reset: automatic or FLOW-BUS","filter":"","id":312,"name":"RstAlarmEn","parameter":156,"value":12},
{"description":"reset: automatic or FLOW-BUS or keyboard","filter":"","id":313,"name":"RstAlarmEn","parameter":156,"value":13},
{"description":"reset: automatic or FLOW-BUS or external","filter":"","id":314,"name":"RstAlarmEn","parameter":156,"value":14},
{"description":"reset: automatic or FLOW-BUS or keyboard or external","filter":"","id":315,"name":"RstAlarmEn","parameter":156,"value":15},
{"description":"no reset possible","filter":"","id":334,"name":"RstCountEn","parameter":157,"value":0},
{"description":"reset: keyboard","filter":"","id":335,"name":"RstCountEn","parameter":157,"value":1},
{"description":"reset: external","filter":"","id":336,"name":"RstCountEn","parameter":157,"value":2},
{"description":"reset: keyboard or external","filter":"","id":337,"name":"RstCountEn","parameter":157,"value":3},
{"description":"reset: FLOW-BUS","filter":"","id":338,"name":"RstCountEn","parameter":157,"value":4},
{"description":"reset: FLOW-BUS or keyboard","filter":"","id":339,"name":"RstCountEn","parameter":157,"value":5},
{"description":"reset: FLOW-BUS or external","filter":"","id":340,"name":"RstCountEn","parameter":157,"value":6},
{"description":"reset: FLOW-BUS or keyboard or external","filter":"","id":341,"name":"RstCountEn","parameter":157,"value":7},
{"description":"always busmaster","filter":"","id":342,"name":"arbitrage","parameter":6,"value":2},
{"description":"enable","filter":"","id":343,"name":"AlarmEnble","parameter":69,"value":1},
{"description":"AD channel 2","filter":"","id":344,"name":"ChanSelect","parameter":71,"value":2},
{"description":"ADC gain = 1x","filter":"&H1C0000","id":356,"name":"ADCcntrReg","parameter":67,"value":0},
{"description":"ADC gain = 2x","filter":"&H1C0000","id":357,"name":"ADCcntrReg","parameter":67,"value":1},
{"description":"ADC gain = 4x","filter":"&H1C0000","id":358,"name":"ADCcntrReg","parameter":67,"value":2},
{"description":"ADC gain = 8x","filter":"&H1C0000","id":359,"name":"ADCcntrReg","parameter":67,"value":3},
{"description":"ADC gain = 16x","filter":"&H1C0000","id":360,"name":"ADCcntrReg","parameter":67,"value":4},
{"description":"ADC gain = 32x","filter":"&H1C0000","id":361,"name":"ADCcntrReg","parameter":67,"value":5},
{"description":"ADC gain = 64x","filter":"&H1C0000","id":362,"name":"ADCcntrReg","parameter":67,"value":6},
{"description":"ADC gain = 128x","filter":"&H1C0000","id":363,"name":"ADCcntrReg","parameter":67,"value":7},
{"description":"ADC bipolar mode","filter":"&H001000","id":374,"name":"ADCcntrReg","parameter":67,"value":0},
{"description":"ADC unipolar mode","filter":"&H001000","id":375,"name":"ADCcntrReg","parameter":67,"value":1},
{"description":"setpoint = master output(FLOW-BUS) * slave factor(analog in)","filter":"","id":377,"name":"cntrlmode","parameter":12,"value":13},
{"description":"no warning message in alarm warning status register","filter":"&H02","id":378,"name":"alarminfo","parameter":28,"value":0},
{"description":"no minimum alarm message (measure>minimum limit)","filter":"&H04","id":379,"name":"alarminfo","parameter":28,"value":0},
{"description":"no maximum alarm message (measure<maximum limit)","filter":"&H08","id":380,"name":"alarminfo","parameter":28,"value":0},
{"description":"batch counter has not reached its limit","filter":"&H10","id":381,"name":"alarminfo","parameter":28,"value":0},
{"description":"response O.K. (setpoint-measure within limit)","filter":"&H20","id":382,"name":"alarminfo","parameter":28,"value":0},
{"description":"master output signal O.K. (or not used)","filter":"&H40","id":383,"name":"alarminfo","parameter":28,"value":0},
{"description":"hardware O.K.","filter":"&H80","id":384,"name":"alarminfo","parameter":28,"value":0},
{"description":"at least 1 error message in alarm error status register","filter":"&H01","id":385,"name":"alarminfo","parameter":28,"value":1},
{"description":"at least 1 warning message in alarm warning status register","filter":"&H02","id":386,"name":"alarminfo","parameter":28,"value":1},
{"description":"minimum alarm message for measured signal","filter":"&H04","id":387,"name":"alarminfo","parameter":28,"value":1},
{"description":"maximum alarm message for measured signal","filter":"&H08","id":388,"name":"alarminfo","parameter":28,"value":1},
{"description":"batch counter has reached its limit","filter":"&H10","id":389,"name":"alarminfo","parameter":28,"value":1},
{"description":"response alarm message: setpoint-measure is too high","filter":"&H20","id":390,"name":"alarminfo","parameter":28,"value":1},
{"description":"master output signal not received: check master instrument","filter":"&H40","id":391,"name":"alarminfo","parameter":28,"value":1},
{"description":"hardware error message: check your hardware","filter":"&H80","id":392,"name":"alarminfo","parameter":28,"value":1},
{"description":"valve in normal position after startup","filter":"&H01","id":393,"name":"ContrType","parameter":166,"value":0},
{"description":"valve in safe position after startup","filter":"&H01","id":394,"name":"ContrType","parameter":166,"value":1},
{"description":"open from zero with PID output to valve","filter":"&H02","id":395,"name":"ContrType","parameter":166,"value":0},
{"description":"open from zero with ramp output to valve","filter":"&H02","id":396,"name":"ContrType","parameter":166,"value":1},
{"description":"fixed monitor output signal","filter":"&H04","id":397,"name":"ContrType","parameter":166,"value":0},
{"description":"monitor output changed at setpoint steps","filter":"&H04","id":398,"name":"ContrType","parameter":166,"value":1},
{"description":"delta-P -5...0 \"W.C.","filter":"","id":399,"name":"PressSensr","parameter":106,"value":4},
{"description":"delta-P -10...0 \"W.C.","filter":"","id":400,"name":"PressSensr","parameter":106,"value":5},
{"description":"delta-P -10...+10 \"W.C.","filter":"","id":401,"name":"PressSensr","parameter":106,"value":6},
{"description":"delta-P 0...1 PSI","filter":"","id":402,"name":"PressSensr","parameter":106,"value":7},
{"description":"delta-P -1...0 PSI","filter":"","id":403,"name":"PressSensr","parameter":106,"value":8},
{"description":"RS232/FLOW-BUS interface","filter":"","id":405,"name":"IdentNr","parameter":175,"value":1},
{"description":"DSCM-E: evaporator controller module (single channel)","filter":"","id":406,"name":"IdentNr","parameter":175,"value":20},
{"description":"DSCM-C: digital single channel module for calibrators","filter":"","id":407,"name":"IdentNr","parameter":175,"value":21},
{"description":"DDCM-A: digital dual channel module for analog instruments","filter":"","id":408,"name":"IdentNr","parameter":175,"value":22},
{"description":"DMCM-D: digital multi channel module for digital instruments","filter":"","id":409,"name":"IdentNr","parameter":175,"value":23},
{"description":"PID controller","filter":"","id":410,"name":"TuningMode","parameter":79,"value":5},
{"description":"Reset module (soft reset)","filter":"","id":413,"name":"Reset","parameter":114,"value":5},
{"description":"voltage drift compensation for valve output turned on","filter":"&H08","id":414,"name":"ContrType","parameter":166,"value":0},
{"description":"voltage drift compensation for valve output turned off","filter":"&H08","id":415,"name":"ContrType","parameter":166,"value":1},
{"description":"alarm when instrument powers-up (eg. after power-down)","filter":"","id":416,"name":"AlrmMode","parameter":118,"value":3},
{"description":"automatic busmaster","filter":"","id":417,"name":"arbitrage","parameter":6,"value":3},
{"description":"auto busmaster and auto bus optimalization (fast token ring)","filter":"","id":418,"name":"arbitrage","parameter":6,"value":67},
{"description":"reset: automatic","filter":"","id":419,"name":"RstCountEn","parameter":157,"value":8},
{"description":"reset: automatic or keyboard","filter":"","id":420,"name":"RstCountEn","parameter":157,"value":9},
{"description":"reset: automatic or external","filter":"","id":421,"name":"RstCountEn","parameter":157,"value":10},
{"description":"reset: automatic or keyboard or external","filter":"","id":422,"name":"RstCountEn","parameter":157,"value":11},
{"description":"reset: automatic or FLOW-BUS","filter":"","id":423,"name":"RstCountEn","parameter":157,"value":12},
{"description":"reset: automatic or FLOW-BUS or keyboard","filter":"","id":424,"name":"RstCountEn","parameter":157,"value":13},
{"description":"reset: automatic or FLOW-BUS or external","filter":"","id":425,"name":"RstCountEn","parameter":157,"value":14},
{"description":"reset: automatic or FLOW-BUS or keyboard or external","filter":"","id":426,"name":"RstCountEn","parameter":157,"value":15},
{"description":"auto slope disabled","filter":"&H10","id":427,"name":"ContrType","parameter":166,"value":0},
{"description":"auto slope enabled for pilot valves","filter":"&H10","id":428,"name":"ContrType","parameter":166,"value":1},
{"description":"Unknown","filter":"","id":429,"name":"DeviceFunc","parameter":185,"value":0},
{"description":"Interface","filter":"","id":430,"name":"DeviceFunc","parameter":185,"value":1},
{"description":"ADDA","filter":"","id":431,"name":"DeviceFunc","parameter":185,"value":2},
{"description":"Operator","filter":"","id":432,"name":"DeviceFunc","parameter":185,"value":3},
{"description":"Supervisor (totalizer/alarm)","filter":"","id":433,"name":"DeviceFunc","parameter":185,"value":4},
{"description":"Controller","filter":"","id":434,"name":"DeviceFunc","parameter":185,"value":5},
{"description":"Meter","filter":"","id":435,"name":"DeviceFunc","parameter":185,"value":6},
{"description":"Special","filter":"","id":436,"name":"DeviceFunc","parameter":185,"value":7},
{"description":"(Protocol) converter","filter":"","id":437,"name":"DeviceFunc","parameter":185,"value":8},
{"description":"Automatic capacity setting for optimal resolution","filter":"&H01","id":438,"name":"CalType","parameter":197,"value":0},
{"description":"Manual capacity setting for optimal resolution","filter":"&H01","id":439,"name":"CalType","parameter":197,"value":1},
{"description":"Barometer value input via parameter 107: BaroPress","filter":"&H02","id":440,"name":"CalType","parameter":197,"value":0},
{"description":"Barometer is master; input automatically from master","filter":"&H02","id":441,"name":"CalType","parameter":197,"value":1},
{"description":"Configuration A: 14 ch. Standard parms. with network scan","filter":"","id":442,"name":"InterfConf","parameter":200,"value":0},
{"description":"Configuration B: 14 ch. Standard parms with fixed chan list","filter":"","id":443,"name":"InterfConf","parameter":200,"value":1},
{"description":"Configuration C: 7 ch. Extended  parms with fixed chan list","filter":"","id":444,"name":"InterfConf","parameter":200,"value":2},
{"description":"(FPP) Range select mode","filter":"","id":446,"name":"cntrlmode","parameter":12,"value":14},
{"description":"(FPP) Manual start sensor select, automatic end sensor","filter":"","id":447,"name":"cntrlmode","parameter":12,"value":15},
{"description":"(FPP) Automatic start sensor select, manual end sensor","filter":"","id":448,"name":"cntrlmode","parameter":12,"value":16},
{"description":"(FPP) Automatic start and end sensor","filter":"","id":449,"name":"cntrlmode","parameter":12,"value":17},
{"description":"setpoint = RS232 setpoint","filter":"","id":450,"name":"cntrlmode","parameter":12,"value":18},
{"description":"automatic correction for valve open turned on","filter":"&H20","id":451,"name":"ContrType","parameter":166,"value":0},
{"description":"automatic correction for valve open turned off","filter":"&H20","id":452,"name":"ContrType","parameter":166,"value":1},
{"description":"controller special mode (valve output steps) turned off","filter":"&H40","id":453,"name":"ContrType","parameter":166,"value":0},
{"description":"controller special mode (valve output steps) turned on","filter":"&H40","id":454,"name":"ContrType","parameter":166,"value":1},
{"description":"Profibus-DP/FLOW-BUS interface module","filter":"","id":455,"name":"IdentNr","parameter":175,"value":24},
{"description":"FLOW-BUS Coriolis Meter","filter":"","id":456,"name":"IdentNr","parameter":175,"value":25},
{"description":"FBI: FLOW-BUS Balance Interface","filter":"","id":457,"name":"IdentNr","parameter":175,"value":26},
{"description":"mass flow in kg/min","filter":"","id":458,"name":"monitor","parameter":60,"value":19},
{"description":"Configuration D: 11 ch. Extended parms with network scan","filter":"","id":459,"name":"InterfConf","parameter":200,"value":3},
{"description":"Analog input enabled","filter":"&H40","id":460,"name":"AnalogMode","parameter":53,"value":0},
{"description":"Analog input disabled","filter":"&H40","id":461,"name":"AnalogMode","parameter":53,"value":1},
{"description":"Analog output enabled","filter":"&H80","id":462,"name":"AnalogMode","parameter":53,"value":0},
{"description":"Analog output disabled","filter":"&H80","id":463,"name":"AnalogMode","parameter":53,"value":1},
{"description":"Diagnostics available in warning register","filter":"&H8000000000000000","id":464,"name":"AlarmReg1","parameter":61,"value":1},
{"description":"No diagnostics available in warning register","filter":"&H8000000000000000","id":466,"name":"AlarmReg1","parameter":61,"value":0},
{"description":"Diagnostics available in error register","filter":"&H8000000000000000","id":468,"name":"AlarmReg2","parameter":62,"value":1},
{"description":"No diagnostics available in error register","filter":"&H8000000000000000","id":470,"name":"AlarmReg2","parameter":62,"value":0},
{"description":"Debug mode off","filter":"","id":472,"name":"DiagMode","parameter":212,"value":0},
{"description":"Debug mode on","filter":"","id":473,"name":"DiagMode","parameter":212,"value":1},
{"description":"set status bit (range 0�127)","filter":"","id":474,"name":"MStatEnabl","parameter":213,"value":0},
{"description":"set status bit (range 0�127)","filter":"","id":475,"name":"MStatEnabl","parameter":213,"value":127},
{"description":"clear all status bits","filter":"","id":476,"name":"MStatEnabl","parameter":213,"value":254},
{"description":"set all status bits","filter":"","id":477,"name":"MStatEnabl","parameter":213,"value":255},
{"description":"No diagnostics available in manufacturer warning register","filter":"&H800000","id":480,"name":"Mwarning","parameter":209,"value":0},
{"description":"Diagnostics available in manufacturer warning register","filter":"&H800000","id":481,"name":"Mwarning","parameter":209,"value":1},
{"description":"No diagnostics available in manufacturer status register","filter":"&H800000","id":484,"name":"Mstatus","parameter":208,"value":0},
{"description":"Diagnostics available in manufacturer status register","filter":"&H800000","id":485,"name":"Mstatus","parameter":208,"value":1},
{"description":"No diagnostics available in manufacturer error register","filter":"&H800000","id":488,"name":"Merror","parameter":210,"value":0},
{"description":"Diagnostics available in manufacturer error register","filter":"&H800000","id":489,"name":"Merror","parameter":210,"value":1},
{"description":"CORIFC: CoriFlow Controller","filter":"","id":490,"name":"IdentNr","parameter":175,"value":27},
{"description":"CORIFM: CoriFlow Meter","filter":"","id":491,"name":"IdentNr","parameter":175,"value":28},
{"description":"setpoint change at batch limit enabled","filter":"","id":493,"name":"CntrStpMod","parameter":126,"value":1},
{"description":"factory zero","filter":"","id":494,"name":"CalMode","parameter":58,"value":18},
{"description":"voltage drive mode","filter":"","id":495,"name":"ValveMode","parameter":232,"value":0},
{"description":"current drive mode","filter":"","id":496,"name":"ValveMode","parameter":232,"value":1},
{"description":"FICC: FLOW-BUS Interface Climate Control","filter":"","id":497,"name":"IdentNr","parameter":175,"value":29},
{"description":"IFI: Instrument FLOW-BUS Interface","filter":"","id":498,"name":"IdentNr","parameter":175,"value":30},
{"description":"KFI: Keithley FLOW-BUS Interface","filter":"","id":499,"name":"IdentNr","parameter":175,"value":31},
{"description":"FSI: FLOW-BUS Switch Interface","filter":"","id":500,"name":"IdentNr","parameter":175,"value":32},
{"description":"RS232 broadcast mode","filter":"","id":501,"name":"cntrlmode","parameter":12,"value":19},
{"description":"valve stearing (valve = setpoint)","filter":"","id":502,"name":"cntrlmode","parameter":12,"value":20},
{"description":"analog valve stearing (valve = analog setpoint)","filter":"","id":503,"name":"cntrlmode","parameter":12,"value":21},
{"description":"Fluidset is enabled","filter":"&H01","id":504,"name":"FldSetProp","parameter":238,"value":1},
{"description":"Fluidset is factory set","filter":"&H02","id":505,"name":"FldSetProp","parameter":238,"value":2},
{"description":"Fluidset is calibrated on actual gas","filter":"&H04","id":506,"name":"FldSetProp","parameter":238,"value":4},
{"description":"Fluidset is disabled","filter":"&H01","id":507,"name":"FldSetProp","parameter":238,"value":0},
{"description":"Fluidset is not factory set","filter":"&H02","id":508,"name":"FldSetProp","parameter":238,"value":0},
{"description":"Fluidset is not calibrated on actual gas","filter":"&H04","id":509,"name":"FldSetProp","parameter":238,"value":0},
{"description":"sensor differentiator (setpoint steps are needed!)","filter":"","id":516,"name":"CalMode","parameter":58,"value":19},
{"description":"Error mode (result of previous cal mode)","filter":"","id":517,"name":"CalMode","parameter":58,"value":255},
{"description":"MSCI: Multi-Sensor/Confroller Interface","filter":"","id":518,"name":"IdentNr","parameter":175,"value":33},
{"description":"Enable zero measure threshold","filter":"&H1000000","id":519,"name":"ADCcntrReg","parameter":67,"value":1},
{"description":"Disable zero measure threshold","filter":"&H1000000","id":520,"name":"ADCcntrReg","parameter":67,"value":0},
{"description":"UFO?: Unidentified FLOW-BUS Object","filter":"","id":521,"name":"IdenrNr","parameter":175,"value":0},
{"description":"read analog jumper (use cntrlmode, read/write)","filter":"&H04","id":522,"name":"IOStatus","parameter":86,"value":1},
{"description":"valve overshoot protection turned on","filter":"&H80","id":523,"name":"ContrType","parameter":166,"value":1},
{"description":"valve overshoot protection turned off","filter":"&H80","id":524,"name":"ContrType","parameter":166,"value":0},
{"description":"APP-D: Active Piston Prover","filter":"","id":525,"name":"IdentNr","parameter":175,"value":34},
{"description":"LFI: Leaktester FLOW-BUS Interface","filter":"","id":526,"name":"IdentNr","parameter":175,"value":35},
{"description":"absolute pressure 0-10 bar","filter":"","id":527,"name":"PressSensr","parameter":106,"value":9},
{"description":"Bridge on","filter":"&H01","id":528,"name":"SensBridge","parameter":295,"value":1},
{"description":"3 windings C","filter":"&H02","id":529,"name":"SensBridge","parameter":295,"value":1},
{"description":"3 windings D","filter":"&H04","id":530,"name":"SensBridge","parameter":295,"value":1},
{"description":"Sensor protection enabled","filter":"&H200","id":531,"name":"SensBridge","parameter":295,"value":1},
{"description":"0 mA","filter":"","id":533,"name":"ValveSafe","parameter":301,"value":0},
{"description":"max mA","filter":"","id":534,"name":"ValveSafe","parameter":301,"value":1},
{"description":"Close","filter":"","id":535,"name":"ValveSafe","parameter":301,"value":2},
{"description":"Open","filter":"","id":536,"name":"ValveSafe","parameter":301,"value":3},
{"description":"Idle","filter":"","id":537,"name":"ValveSafe","parameter":301,"value":4},
{"description":"Value (for DeviceNet only)","filter":"","id":538,"name":"ValveSafe","parameter":301,"value":5},
{"description":"Automatic sensor configuration on","filter":"&H100","id":540,"name":"SensBridge","parameter":295,"value":1},
{"description":"FLOW-BUS","filter":"","id":541,"name":"Bus1Select","parameter":305,"value":0},
{"description":"Modbus","filter":"","id":542,"name":"Bus1Select","parameter":305,"value":1},
{"description":"ProPar","filter":"","id":543,"name":"Bus1Select","parameter":305,"value":2},
{"description":"RS232","filter":"","id":544,"name":"Bus1Medium","parameter":306,"value":0},
{"description":"RS485","filter":"","id":545,"name":"Bus1Medium","parameter":306,"value":1},
{"description":"Normal","filter":"","id":546,"name":"Bus2Mode","parameter":307,"value":0},
{"description":"Config mode","filter":"","id":547,"name":"Bus2Mode","parameter":307,"value":1},
{"description":"FLOW-BUS","filter":"","id":549,"name":"Bus2Select","parameter":308,"value":0},
{"description":"Modbus","filter":"","id":550,"name":"Bus2Select","parameter":308,"value":1},
{"description":"ProPar","filter":"","id":551,"name":"Bus2Select","parameter":308,"value":2},
{"description":"RS485","filter":"","id":552,"name":"Bus2Medium","parameter":311,"value":1},
{"description":"RS232","filter":"","id":553,"name":"Bus2Medium","parameter":311,"value":0},
{"description":"Analog input","filter":"","id":554,"name":"PIOChSel","parameter":314,"value":0},
{"description":"Analog output","filter":"","id":555,"name":"PIOChSel","parameter":314,"value":1},
{"description":"General purpose in-/output","filter":"","id":556,"name":"PIOChSel","parameter":314,"value":2},
{"description":"Voltage output (0..10 V)","filter":"","id":557,"name":"PIOCfgSel","parameter":319,"value":0},
{"description":"Current output (0..20 mA)","filter":"","id":558,"name":"PIOCfgSel","parameter":319,"value":1},
{"description":"Digital output","filter":"","id":559,"name":"PIOCfgSel","parameter":319,"value":2},
{"description":"Frequency output","filter":"","id":560,"name":"PIOCfgSel","parameter":319,"value":3},
{"description":"Duty cycle output (20 kHz)","filter":"","id":561,"name":"PIOCfgSel","parameter":319,"value":4},
{"description":"Digital pulse output","filter":"","id":562,"name":"PIOCfgSel","parameter":319,"value":5},
{"description":"Voltage input (0..10 V)","filter":"","id":563,"name":"PIOCfgSel","parameter":319,"value":6},
{"description":"Current input (0..20 mA)","filter":"","id":564,"name":"PIOCfgSel","parameter":319,"value":7},
{"description":"Digital input","filter":"","id":565,"name":"PIOCfgSel","parameter":319,"value":8},
{"description":"Disabled","filter":"","id":566,"name":"PIOCfgSel","parameter":319,"value":255},
{"description":"automatic sensor configuration","filter":"","id":567,"name":"CalMode","parameter":58,"value":20},
{"description":"sensor temperature calibration","filter":"","id":568,"name":"CalMode","parameter":58,"value":21},
{"description":"customer zero (no control mode 9 needed)","filter":"","id":569,"name":"CalMode","parameter":58,"value":22},
{"description":"valve safe state","filter":"","id":570,"name":"cntrlmode","parameter":12,"value":22},
{"description":"PROFIBUS DP","filter":"","id":571,"name":"Bus1Select","parameter":305,"value":13},
{"description":"DeviceNet","filter":"","id":572,"name":"Bus1Select","parameter":305,"value":10},
{"description":"NoBus","filter":"","id":573,"name":"Bus1Select","parameter":305,"value":255},
{"description":"Setpoint","filter":"","id":574,"name":"SetMonitor","parameter":329,"value":0},
{"description":"Filtered setpoint","filter":"","id":575,"name":"SetMonitor","parameter":329,"value":1},
{"description":"Setpoint after linear slope","filter":"","id":576,"name":"SetMonitor","parameter":329,"value":2},
{"description":"None","filter":"","id":577,"name":"Bus1Parity","parameter":335,"value":0},
{"description":"Odd","filter":"","id":578,"name":"Bus1Parity","parameter":335,"value":1},
{"description":"Even","filter":"","id":579,"name":"Bus1Parity","parameter":335,"value":2},
{"description":"None","filter":"","id":580,"name":"Bus2Parity","parameter":336,"value":0},
{"description":"Odd","filter":"","id":581,"name":"Bus2Parity","parameter":336,"value":1},
{"description":"Even","filter":"","id":582,"name":"Bus2Parity","parameter":336,"value":2},
{"description":"0V (for PIO use)","filter":"","id":583,"name":"ValveMode","parameter":232,"value":2},
{"description":"auto busmaster and auto bus optimalization with fail safe","filter":"","id":584,"name":"arbitrage","parameter":6,"value":83},
{"description":"Main PCB serial number","filter":"","id":585,"name":"PCBSerialI","parameter":340,"value":0},
{"description":"Sensor PCB serial number","filter":"","id":586,"name":"PCBSerialI","parameter":340,"value":1},
{"description":"temperature (controller)","filter":"","id":588,"name":"sensortype","parameter":22,"value":5},
{"description":"temperature (sensor)","filter":"","id":589,"name":"sensortype","parameter":22,"value":133},
{"description":"time (duration) (controller)","filter":"","id":590,"name":"sensortype","parameter":22,"value":6},
{"description":"time (durarion) (sensor)","filter":"","id":591,"name":"sensortype","parameter":22,"value":134},
{"description":"frequency (controller)","filter":"","id":592,"name":"sensortype","parameter":22,"value":7},
{"description":"frequency (sensor)","filter":"","id":593,"name":"sensortype","parameter":22,"value":135},
{"description":"weight (controller)","filter":"","id":594,"name":"sensortype","parameter":22,"value":8},
{"description":"weight (sensor)","filter":"","id":595,"name":"sensortype","parameter":22,"value":136},
{"description":"solutions (controller","filter":"","id":596,"name":"sensortype","parameter":22,"value":9},
{"description":"solutions (sensor)","filter":"","id":597,"name":"sensortype","parameter":22,"value":137},
{"description":"Volume fractions","filter":"","id":598,"name":"MixFrcType","parameter":346,"value":0},
{"description":"Mass fractions","filter":"","id":599,"name":"MixFrcType","parameter":346,"value":1},
{"description":"Mole fractions","filter":"","id":600,"name":"MixFrcType","parameter":346,"value":2},
{"description":"Digital Batch Flow Controller","filter":"","id":602,"name":"IdentNr","parameter":175,"value":36},
{"description":"Digital PID Controller","filter":"","id":603,"name":"IdentNr","parameter":175,"value":37},
{"description":"vent DEPx (when available)","filter":"","id":604,"name":"cntrlmode","parameter":12,"value":24},
{"description":"PROFINET","filter":"","id":605,"name":"Bus1Select","parameter":305,"value":14},
{"description":"EtherCAT","filter":"","id":606,"name":"Bus1Select","parameter":305,"value":11},
{"description":"Temperature from fluidset temperature parameter (p181)","filter":"&H01","id":607,"name":"FlConCondS","parameter":354,"value":0},
{"description":"Temperature from temperature sensor parameter (p142)","filter":"&H01","id":608,"name":"FlConCondS","parameter":354,"value":1},
{"description":"Pressure from fluidset pressure inlet parameter (p178)","filter":"&H02","id":609,"name":"FlConCondS","parameter":354,"value":0},
{"description":"Pressure from pressure sensor parameter (p143)","filter":"&H02","id":610,"name":"FlConCondS","parameter":354,"value":2},
{"description":"Density","filter":"","id":611,"name":"ActFlPIdx","parameter":344,"value":0},
{"description":"Heat capacity","filter":"","id":612,"name":"ActFlPIdx","parameter":344,"value":1},
{"description":"Thermal conductivity","filter":"","id":613,"name":"ActFlPIdx","parameter":344,"value":2},
{"description":"Viscosity","filter":"","id":614,"name":"ActFlPIdx","parameter":344,"value":3},
{"description":"Use fluidset conditions for conversion (instead of p354)","filter":"","id":616,"name":"cntrlmode","parameter":12,"value":23},
{"description":"Bus Gateway","filter":"","id":617,"name":"IdentNr","parameter":175,"value":38},
{"description":"DTC: digital temperature controller","filter":"","id":618,"name":"IdentNr","parameter":175,"value":39},
{"description":"DTM: digital temperature meter","filter":"","id":619,"name":"IdentNr","parameter":175,"value":40},
{"description":"density (controller)","filter":"","id":620,"name":"sensortype","parameter":22,"value":10},
{"description":"density (sensor)","filter":"","id":621,"name":"sensortype","parameter":22,"value":138},
{"description":"Fieldbus interface (main interface, fieldbus1)","filter":"","id":622,"name":"BusIntIdx","parameter":378,"value":0},
{"description":"Instrument interface (fieldbus2)","filter":"","id":623,"name":"BusIntIdx","parameter":378,"value":1},
{"description":"Display interface","filter":"","id":624,"name":"BusIntIdx","parameter":378,"value":2},
{"description":"Service interface","filter":"","id":625,"name":"BusIntIdx","parameter":378,"value":3},
{"description":"HART interface","filter":"","id":626,"name":"BusIntIdx","parameter":378,"value":4},
{"description":"DC: digital controller","filter":"","id":627,"name":"IdentNr","parameter":175,"value":41},
{"description":"DM: digital meter","filter":"","id":628,"name":"IdentNr","parameter":175,"value":42},
{"description":"Production test completed succesfully","filter":"&H01","id":629,"name":"SelfStat","parameter":379,"value":1},
{"description":"Service test completed","filter":"&H02","id":630,"name":"SelfStat","parameter":379,"value":2},
{"description":"Production test detection enabled","filter":"&H04","id":631,"name":"SelfStat","parameter":379,"value":4},
{"description":"Service test detection enabled","filter":"&H08","id":632,"name":"SelfStat","parameter":379,"value":8},
{"description":"Continuous","filter":"","id":633,"name":"DosType","parameter":398,"value":0},
{"description":"Ratio","filter":"","id":634,"name":"DosType","parameter":398,"value":1},
{"description":"Batch","filter":"","id":635,"name":"DosType","parameter":398,"value":2},
{"description":"PID controlled","filter":"","id":636,"name":"DosCtrType","parameter":399,"value":0},
{"description":"On/off controlled","filter":"","id":637,"name":"DosCtrType","parameter":399,"value":1},
{"description":"Pump/control valve","filter":"","id":638,"name":"DosActType","parameter":400,"value":0},
{"description":"On/off valve","filter":"","id":639,"name":"DosActType","parameter":400,"value":1},
{"description":"External trigger [dosing type=batch]","filter":"","id":640,"name":"DosStartTp","parameter":401,"value":0},
{"description":"Time interval [dosing type=batch]","filter":"","id":641,"name":"DosStartTp","parameter":401,"value":1}]}